-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Dec 23 10:36:19 2021 UTC
-- File 20210823_135928_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.662245 to 32.231813
-each separately from 8s to 32s and from 21894us to 662534us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.359360 in 1 packets: 18
      14.209341 in 1 packets: 36
MIU packet time from 2021-08-23T13:59:39.662.245.000 to 2021-08-23T14:00:03.231.813.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.359360 in 1 packets: 18
      14.209341 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.8410304 to 33.8409509
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999669 in 1 packets: 18
      13.9999536 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210823_135928_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.5032761 to 10.5035651
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.3593270  -- OK for SL --
   MIU time from 2021-08-23T13:59:39.662.245.000 to 2021-08-23T13:59:39.662.534.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.359360

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.8628921 to 19.8631811
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.2092950  -- OK for SL --
   MIU time from 2021-08-23T13:59:49.021.894.000 to 2021-08-23T13:59:49.022.183.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.209341

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.0724761 to 34.0727651
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-23T14:00:03.231.524.000 to 2021-08-23T14:00:03.231.813.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.503276100 corrected by 40bit 10MHz counter of 9.841030400 obtained at 1Hz counter of 8. Result: 8.662247819
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-08-23T13:59:39.662.104.725
   Max 1st sample MIU time among all SL packets: 2021-08-23T14:00:03.231.672.725
   Min 1st sample XO time among all SL packets: 8.662107544
   Max 1st sample XO time among all SL packets: 32.231674666
   TM values from -272 to 231 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-23T13:59:39.662.104.725 (XO: 8.662107544 - 8.662413545, t=37.2 degC)  Rank: 00081
   TM values: Max =   220  Min =  -257  Ave =    -4.50  Std =    21.47
   Min TM value:-257 at an offset of 120.538 usec -> 2021-08-23T13:59:39.662.225.262 (XO: 8.662228082)
   Max TM value:220 at an offset of 117.988 usec -> 2021-08-23T13:59:39.662.222.712 (XO: 8.662225532)
   Max dTM/dt [1/us]:2.60e+03 at an offset of 120.650 usec -> 2021-08-23T13:59:39.662.225.375 (XO: 8.662228194)
   Min dTM/dt [1/us]:-2.47e+03 at an offset of 118.100 usec -> 2021-08-23T13:59:39.662.222.825 (XO: 8.662225644)

 - Block number: 2 starting at 2021-08-23T13:59:49.021.753.725 (XO: 18.021754597 - 18.022060598, t=37.2 degC)  Rank: 00083
   TM values: Max =   226  Min =  -260  Ave =    -4.54  Std =    22.52
   Min TM value:-260 at an offset of 108.950 usec -> 2021-08-23T13:59:49.021.862.675 (XO: 18.021863547)
   Max TM value:226 at an offset of 108.688 usec -> 2021-08-23T13:59:49.021.862.412 (XO: 18.021863285)
   Max dTM/dt [1/us]:2.37e+03 at an offset of 111.350 usec -> 2021-08-23T13:59:49.021.865.075 (XO: 18.021865947)
   Min dTM/dt [1/us]:-2.58e+03 at an offset of 108.800 usec -> 2021-08-23T13:59:49.021.862.525 (XO: 18.021863397)

 - Block number: 3 starting at 2021-08-23T14:00:03.231.383.725 (XO: 32.231385665 - 32.231691666, t=37.2 degC)  Rank: 00080
   TM values: Max =   231  Min =  -272  Ave =    -4.52  Std =    22.40
   Min TM value:-272 at an offset of 108.100 usec -> 2021-08-23T14:00:03.231.491.825 (XO: 32.231493766)
   Max TM value:231 at an offset of 106.038 usec -> 2021-08-23T14:00:03.231.489.762 (XO: 32.231491703)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 105.900 usec -> 2021-08-23T14:00:03.231.489.625 (XO: 32.231491566)
   Min dTM/dt [1/us]:-2.22e+03 at an offset of 107.963 usec -> 2021-08-23T14:00:03.231.491.687 (XO: 32.231493628)
