-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Dec 23 08:18:18 2021 UTC
-- File 20210823_121741_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.992341 to 39.261580
-each separately from 8s to 39s and from 261291us to 992630us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.759370 in 1 packets: 18
      22.509002 in 1 packets: 36
MIU packet time from 2021-08-23T12:17:52.992.341.000 to 2021-08-23T12:18:23.261.580.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.759370 in 1 packets: 18
      22.509002 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.8490745 to 40.8489720
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999735 in 1 packets: 18
      22.9999240 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210823_121741_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.8414159 to 10.8417049
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.7593430  -- OK for SL --
   MIU time from 2021-08-23T12:17:52.992.341.000 to 2021-08-23T12:17:52.992.630.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.759370

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.6010479 to 18.6013369
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.5089270  -- OK for SL --
   MIU time from 2021-08-23T12:18:00.752.000.000 to 2021-08-23T12:18:00.752.289.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.509002

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.1102639 to 41.1105529
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-23T12:18:23.261.291.000 to 2021-08-23T12:18:23.261.580.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.841415900 corrected by 40bit 10MHz counter of 9.849074500 obtained at 1Hz counter of 8. Result: 8.992344675
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-08-23T12:17:52.992.200.725
   Max 1st sample MIU time among all SL packets: 2021-08-23T12:18:23.261.439.725
   Min 1st sample XO time among all SL packets: 8.992204399
   Max 1st sample XO time among all SL packets: 39.261441462
   TM values from -261 to 215 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-23T12:17:52.992.200.725 (XO: 8.992204399 - 8.992510400, t=37.2 degC)  Rank: 00079
   TM values: Max =   215  Min =  -257  Ave =    -4.41  Std =    21.14
   Min TM value:-257 at an offset of 126.650 usec -> 2021-08-23T12:17:52.992.327.375 (XO: 8.992331050)
   Max TM value:215 at an offset of 126.363 usec -> 2021-08-23T12:17:52.992.327.087 (XO: 8.992330762)
   Max dTM/dt [1/us]:2.13e+03 at an offset of 129.025 usec -> 2021-08-23T12:17:52.992.329.750 (XO: 8.992333425)
   Min dTM/dt [1/us]:-2.20e+03 at an offset of 126.463 usec -> 2021-08-23T12:17:52.992.327.187 (XO: 8.992330862)

 - Block number: 2 starting at 2021-08-23T12:18:00.751.859.725 (XO: 16.751862106 - 16.752168107, t=37.2 degC)  Rank: 00079
   TM values: Max =   192  Min =  -261  Ave =    -4.51  Std =    21.21
   Min TM value:-261 at an offset of 127.825 usec -> 2021-08-23T12:18:00.751.987.550 (XO: 16.751989932)
   Max TM value:192 at an offset of 127.513 usec -> 2021-08-23T12:18:00.751.987.237 (XO: 16.751989619)
   Max dTM/dt [1/us]:1.71e+03 at an offset of 130.225 usec -> 2021-08-23T12:18:00.751.989.950 (XO: 16.751992332)
   Min dTM/dt [1/us]:-1.56e+03 at an offset of 127.625 usec -> 2021-08-23T12:18:00.751.987.350 (XO: 16.751989732)

 - Block number: 3 starting at 2021-08-23T12:18:23.261.150.725 (XO: 39.261152461 - 39.261458462, t=37.2 degC)  Rank: 00083
   TM values: Max =   215  Min =  -232  Ave =    -4.44  Std =    22.39
   Min TM value:-232 at an offset of 127.025 usec -> 2021-08-23T12:18:23.261.277.750 (XO: 39.261279486)
   Max TM value:215 at an offset of 129.713 usec -> 2021-08-23T12:18:23.261.280.437 (XO: 39.261282174)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 129.600 usec -> 2021-08-23T12:18:23.261.280.325 (XO: 39.261282061)
   Min dTM/dt [1/us]:-1.86e+03 at an offset of 126.838 usec -> 2021-08-23T12:18:23.261.277.562 (XO: 39.261279299)
