-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Dec 23 06:32:13 2021 UTC
-- File 20210823_111020_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.232417 to 69.745329
-each separately from 11s to 69s and from 86776us to 745329us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.978259 in 1 packets: 18
      8.278633 in 1 packets: 36
      26.596600 in 1 packets: 54
      12.657975 in 1 packets: 72
MIU packet time from 2021-08-23T11:10:34.232.417.000 to 2021-08-23T11:11:32.745.329.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.978259 in 1 packets: 18
      8.278633 in 1 packets: 36
      26.596600 in 1 packets: 54
      12.657975 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.9267758 to 70.9265845
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999638 in 1 packets: 18
      7.9999736 in 1 packets: 36
      26.9999109 in 1 packets: 54
      11.9999604 in 1 packets: 72
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 73 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Number of generated alerts: 
      0 in 73 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210823_111020_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.1591935 to 13.1594825
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.9782230  -- OK for SL --
   MIU time from 2021-08-23T11:10:34.232.417.000 to 2021-08-23T11:10:34.232.706.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.978259

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.1377055 to 24.1379945
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.2786070  -- OK for SL --
   MIU time from 2021-08-23T11:10:45.210.965.000 to 2021-08-23T11:10:45.211.254.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.278633

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.4166015 to 32.4168905
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.5965110  -- OK for SL --
   MIU time from 2021-08-23T11:10:53.489.887.000 to 2021-08-23T11:10:53.490.176.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.596600

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 59.0134015 to 59.0136905
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.6579350  -- OK for SL --
   MIU time from 2021-08-23T11:11:20.086.776.000 to 2021-08-23T11:11:20.087.065.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.657975

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      92 in 18 packets (all packets)  -- OK --
   10 MHz counter from 71.6716255 to 71.6719145
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-23T11:11:32.745.040.000 to 2021-08-23T11:11:32.745.329.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.159193500 corrected by 40bit 10MHz counter of 12.926775800 obtained at 1Hz counter of 11. Result: 11.232418467
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-23T11:10:34.232.276.725
   Max 1st sample MIU time among all SL packets: 2021-08-23T11:11:32.745.188.725
   Min 1st sample XO time among all SL packets: 11.232278192
   Max 1st sample XO time among all SL packets: 69.745192184
   TM values from -261 to 209 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-23T11:10:34.232.276.725 (XO: 11.232278192 - 11.232584193, t=37.2 degC)  Rank: 00076
   TM values: Max =   203  Min =  -244  Ave =    -4.54  Std =    21.80
   Min TM value:-244 at an offset of 119.275 usec -> 2021-08-23T11:10:34.232.396.000 (XO: 11.232397467)
   Max TM value:203 at an offset of 121.775 usec -> 2021-08-23T11:10:34.232.398.500 (XO: 11.232399967)
   Max dTM/dt [1/us]:2.17e+03 at an offset of 121.600 usec -> 2021-08-23T11:10:34.232.398.325 (XO: 11.232399792)
   Min dTM/dt [1/us]:-2.11e+03 at an offset of 119.063 usec -> 2021-08-23T11:10:34.232.395.787 (XO: 11.232397254)

 - Block number: 2 starting at 2021-08-23T11:10:45.210.824.725 (XO: 22.210826321 - 22.211132322, t=37.2 degC)  Rank: 00080
   TM values: Max =   187  Min =  -222  Ave =    -4.52  Std =    18.56
   Min TM value:-222 at an offset of 126.000 usec -> 2021-08-23T11:10:45.210.950.725 (XO: 22.210952321)
   Max TM value:187 at an offset of 128.638 usec -> 2021-08-23T11:10:45.210.953.362 (XO: 22.210954959)
   Max dTM/dt [1/us]:1.46e+03 at an offset of 128.475 usec -> 2021-08-23T11:10:45.210.953.200 (XO: 22.210954796)
   Min dTM/dt [1/us]:-1.73e+03 at an offset of 125.800 usec -> 2021-08-23T11:10:45.210.950.525 (XO: 22.210952121)

 - Block number: 3 starting at 2021-08-23T11:10:53.489.746.725 (XO: 30.489749641 - 30.490055642, t=37.2 degC)  Rank: 00080
   TM values: Max =   209  Min =  -261  Ave =    -4.50  Std =    20.13
   Min TM value:-261 at an offset of 131.663 usec -> 2021-08-23T11:10:53.489.878.387 (XO: 30.489881304)
   Max TM value:209 at an offset of 131.400 usec -> 2021-08-23T11:10:53.489.878.125 (XO: 30.489881042)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 134.063 usec -> 2021-08-23T11:10:53.489.880.787 (XO: 30.489883704)
   Min dTM/dt [1/us]:-2.40e+03 at an offset of 131.513 usec -> 2021-08-23T11:10:53.489.878.237 (XO: 30.489881154)

 - Block number: 4 starting at 2021-08-23T11:11:20.086.635.725 (XO: 57.086637411 - 57.086943412, t=37.2 degC)  Rank: 00077
   TM values: Max =   186  Min =  -230  Ave =    -4.52  Std =    18.16
   Min TM value:-230 at an offset of 116.713 usec -> 2021-08-23T11:11:20.086.752.437 (XO: 57.086754124)
   Max TM value:186 at an offset of 116.450 usec -> 2021-08-23T11:11:20.086.752.175 (XO: 57.086753861)
   Max dTM/dt [1/us]:2.11e+03 at an offset of 119.125 usec -> 2021-08-23T11:11:20.086.754.850 (XO: 57.086756536)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 116.550 usec -> 2021-08-23T11:11:20.086.752.275 (XO: 57.086753961)

 - Block number: 5 starting at 2021-08-23T11:11:32.744.899.725 (XO: 69.744903183 - 69.745209184, t=37.2 degC)  Rank: 00092
   TM values: Max =   192  Min =  -249  Ave =    -4.55  Std =    20.33
   Min TM value:-249 at an offset of 125.575 usec -> 2021-08-23T11:11:32.745.025.300 (XO: 69.745028759)
   Max TM value:192 at an offset of 128.150 usec -> 2021-08-23T11:11:32.745.027.875 (XO: 69.745031334)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 127.950 usec -> 2021-08-23T11:11:32.745.027.675 (XO: 69.745031134)
   Min dTM/dt [1/us]:-1.93e+03 at an offset of 125.363 usec -> 2021-08-23T11:11:32.745.025.087 (XO: 69.745028546)
