-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec 22 15:53:15 2021 UTC
-- File 20210822_133833_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.294337 to 28.344615
-each separately from 7s to 28s and from 294337us to 344615us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.049700 in 1 packets: 18
MIU packet time from 2021-08-22T13:38:43.294.337.000 to 2021-08-22T13:39:04.344.615.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.049700 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.8716831 to 29.8716138
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999307 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210822_133833_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.1660209 to 9.1663099
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.0496310  -- OK for SL --
   MIU time from 2021-08-22T13:38:43.294.337.000 to 2021-08-22T13:38:43.294.626.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.049700

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.2159409 to 30.2162299
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-22T13:39:04.344.326.000 to 2021-08-22T13:39:04.344.615.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.166020900 corrected by 40bit 10MHz counter of 8.871683100 obtained at 1Hz counter of 7. Result: 7.294338771
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-22T13:38:43.294.196.725
   Max 1st sample MIU time among all SL packets: 2021-08-22T13:39:04.344.474.725
   Min 1st sample XO time among all SL packets: 7.294198496
   Max 1st sample XO time among all SL packets: 28.344476962
   TM values from -253 to 218 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-22T13:38:43.294.196.725 (XO: 7.294198496 - 7.294504497, t=37.2 degC)  Rank: 00078
   TM values: Max =   212  Min =  -253  Ave =    -4.54  Std =    18.66
   Min TM value:-253 at an offset of 121.763 usec -> 2021-08-22T13:38:43.294.318.487 (XO: 7.294320259)
   Max TM value:212 at an offset of 121.463 usec -> 2021-08-22T13:38:43.294.318.187 (XO: 7.294319959)
   Max dTM/dt [1/us]:2.24e+03 at an offset of 124.113 usec -> 2021-08-22T13:38:43.294.320.837 (XO: 7.294322609)
   Min dTM/dt [1/us]:-2.19e+03 at an offset of 121.563 usec -> 2021-08-22T13:38:43.294.318.287 (XO: 7.294320059)

 - Block number: 2 starting at 2021-08-22T13:39:04.344.185.725 (XO: 28.344187961 - 28.344493962, t=37.2 degC)  Rank: 00081
   TM values: Max =   218  Min =  -210  Ave =    -4.57  Std =    24.58
   Min TM value:-210 at an offset of 106.300 usec -> 2021-08-22T13:39:04.344.292.025 (XO: 28.344294261)
   Max TM value:218 at an offset of 175.926 usec -> 2021-08-22T13:39:04.344.361.650 (XO: 28.344363886)
   Max dTM/dt [1/us]:2.11e+03 at an offset of 175.801 usec -> 2021-08-22T13:39:04.344.361.525 (XO: 28.344363761)
   Min dTM/dt [1/us]:-1.64e+03 at an offset of 106.150 usec -> 2021-08-22T13:39:04.344.291.875 (XO: 28.344294111)
