-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec 22 12:32:21 2021 UTC
-- File 20210822_113932_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.685269 to 19.414374
-each separately from 9s to 19s and from 414085us to 685558us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.728527 in 1 packets: 18
MIU packet time from 2021-08-22T11:39:44.685.269.000 to 2021-08-22T11:39:54.414.374.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.728527 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.9148066 to 20.9147736
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999670 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210822_113932_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.6000760 to 11.6003650
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.7284950  -- OK for SL --
   MIU time from 2021-08-22T11:39:44.685.269.000 to 2021-08-22T11:39:44.685.558.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.728527

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.3288600 to 21.3291490
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-22T11:39:54.414.085.000 to 2021-08-22T11:39:54.414.374.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.600076000 corrected by 40bit 10MHz counter of 10.914806600 obtained at 1Hz counter of 9. Result: 9.685271730
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-22T11:39:44.685.128.725
   Max 1st sample MIU time among all SL packets: 2021-08-22T11:39:54.414.233.725
   Min 1st sample XO time among all SL packets: 9.685131454
   Max 1st sample XO time among all SL packets: 19.414236492
   TM values from -238 to 221 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-22T11:39:44.685.128.725 (XO: 9.685131454 - 9.685437455, t=37.2 degC)  Rank: 00083
   TM values: Max =   199  Min =  -238  Ave =    -4.59  Std =    16.75
   Min TM value:-238 at an offset of 121.563 usec -> 2021-08-22T11:39:44.685.250.287 (XO: 9.685253017)
   Max TM value:199 at an offset of 124.138 usec -> 2021-08-22T11:39:44.685.252.862 (XO: 9.685255592)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 123.950 usec -> 2021-08-22T11:39:44.685.252.675 (XO: 9.685255405)
   Min dTM/dt [1/us]:-1.93e+03 at an offset of 121.363 usec -> 2021-08-22T11:39:44.685.250.087 (XO: 9.685252817)

 - Block number: 2 starting at 2021-08-22T11:39:54.413.944.725 (XO: 19.413947491 - 19.414253492, t=37.2 degC)  Rank: 00086
   TM values: Max =   221  Min =  -237  Ave =    -4.52  Std =    19.13
   Min TM value:-237 at an offset of 122.788 usec -> 2021-08-22T11:39:54.414.067.512 (XO: 19.414070279)
   Max TM value:221 at an offset of 122.500 usec -> 2021-08-22T11:39:54.414.067.225 (XO: 19.414069991)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 125.138 usec -> 2021-08-22T11:39:54.414.069.862 (XO: 19.414072629)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 122.613 usec -> 2021-08-22T11:39:54.414.067.337 (XO: 19.414070104)
