-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec 22 11:54:30 2021 UTC
-- File 20210822_112241_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.449790 to 41.609122
-each separately from 1s to 41s and from 319337us to 609122us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.869258 in 1 packets: 18
      22.289207 in 1 packets: 36
MIU packet time from 2021-08-22T11:22:45.449.790.000 to 2021-08-22T11:23:25.609.122.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.869258 in 1 packets: 18
      22.289207 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.9823681 to 42.9822364
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999408 in 1 packets: 18
      21.9999275 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210822_112241_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.4321586 to 3.4324476
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.8691990  -- OK for SL --
   MIU time from 2021-08-22T11:22:45.449.790.000 to 2021-08-22T11:22:45.450.079.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.869258

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.3016466 to 21.3019356
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.2891350  -- OK for SL --
   MIU time from 2021-08-22T11:23:03.319.337.000 to 2021-08-22T11:23:03.319.626.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.289207

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      93 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.5910706 to 43.5913596
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-22T11:23:25.608.833.000 to 2021-08-22T11:23:25.609.122.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.432158600 corrected by 40bit 10MHz counter of 2.982368100 obtained at 1Hz counter of 1. Result: 1.449791984
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-08-22T11:22:45.449.649.725
   Max 1st sample MIU time among all SL packets: 2021-08-22T11:23:25.608.981.725
   Min 1st sample XO time among all SL packets: 1.449651709
   Max 1st sample XO time among all SL packets: 41.608984874
   TM values from -244 to 202 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-22T11:22:45.449.649.725 (XO: 1.449651709 - 1.449957710, t=37.2 degC)  Rank: 00081
   TM values: Max =   190  Min =  -230  Ave =    -4.53  Std =    18.65
   Min TM value:-230 at an offset of 129.800 usec -> 2021-08-22T11:22:45.449.779.525 (XO: 1.449781509)
   Max TM value:190 at an offset of 132.413 usec -> 2021-08-22T11:22:45.449.782.137 (XO: 1.449784122)
   Max dTM/dt [1/us]:1.53e+03 at an offset of 132.225 usec -> 2021-08-22T11:22:45.449.781.950 (XO: 1.449783934)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 129.613 usec -> 2021-08-22T11:22:45.449.779.337 (XO: 1.449781322)

 - Block number: 2 starting at 2021-08-22T11:23:03.319.196.725 (XO: 19.319198510 - 19.319504511, t=37.2 degC)  Rank: 00080
   TM values: Max =   202  Min =  -244  Ave =    -4.54  Std =    20.39
   Min TM value:-244 at an offset of 128.888 usec -> 2021-08-22T11:23:03.319.325.612 (XO: 19.319327398)
   Max TM value:202 at an offset of 131.425 usec -> 2021-08-22T11:23:03.319.328.150 (XO: 19.319329936)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 131.275 usec -> 2021-08-22T11:23:03.319.328.000 (XO: 19.319329786)
   Min dTM/dt [1/us]:-1.97e+03 at an offset of 128.738 usec -> 2021-08-22T11:23:03.319.325.462 (XO: 19.319327248)

 - Block number: 3 starting at 2021-08-22T11:23:25.608.692.725 (XO: 41.608695873 - 41.609001874, t=37.2 degC)  Rank: 00093
   TM values: Max =   190  Min =  -243  Ave =    -4.53  Std =    20.66
   Min TM value:-243 at an offset of 129.200 usec -> 2021-08-22T11:23:25.608.821.925 (XO: 41.608825073)
   Max TM value:190 at an offset of 131.788 usec -> 2021-08-22T11:23:25.608.824.512 (XO: 41.608827661)
   Max dTM/dt [1/us]:1.77e+03 at an offset of 131.575 usec -> 2021-08-22T11:23:25.608.824.300 (XO: 41.608827448)
   Min dTM/dt [1/us]:-1.89e+03 at an offset of 129.013 usec -> 2021-08-22T11:23:25.608.821.737 (XO: 41.608824886)
