-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec 22 11:46:55 2021 UTC
-- File 20210822_111903_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.114328 to 28.864748
-each separately from 8s to 28s and from 114328us to 864748us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.749842 in 1 packets: 18
MIU packet time from 2021-08-22T11:19:14.114.328.000 to 2021-08-22T11:19:34.864.748.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.749842 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.9454468 to 29.9453809
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999341 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210822_111903_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      95 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.0597763 to 10.0600653
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.7497750  -- OK for SL --
   MIU time from 2021-08-22T11:19:14.114.328.000 to 2021-08-22T11:19:14.114.617.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.749842

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.8098403 to 30.8101293
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-22T11:19:34.864.459.000 to 2021-08-22T11:19:34.864.748.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.059776300 corrected by 40bit 10MHz counter of 9.945446800 obtained at 1Hz counter of 8. Result: 8.114329877
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-22T11:19:14.114.187.725
   Max 1st sample MIU time among all SL packets: 2021-08-22T11:19:34.864.607.725
   Min 1st sample XO time among all SL packets: 8.114189602
   Max 1st sample XO time among all SL packets: 28.864610978
   TM values from -254 to 208 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-22T11:19:14.114.187.725 (XO: 8.114189602 - 8.114495603, t=37.2 degC)  Rank: 00095
   TM values: Max =   197  Min =  -254  Ave =    -4.53  Std =    21.26
   Min TM value:-254 at an offset of 122.738 usec -> 2021-08-22T11:19:14.114.310.462 (XO: 8.114312340)
   Max TM value:197 at an offset of 120.713 usec -> 2021-08-22T11:19:14.114.308.437 (XO: 8.114310315)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 125.113 usec -> 2021-08-22T11:19:14.114.312.837 (XO: 8.114314715)
   Min dTM/dt [1/us]:-1.83e+03 at an offset of 122.550 usec -> 2021-08-22T11:19:14.114.310.275 (XO: 8.114312152)

 - Block number: 2 starting at 2021-08-22T11:19:34.864.318.725 (XO: 28.864321977 - 28.864627978, t=37.2 degC)  Rank: 00080
   TM values: Max =   208  Min =  -240  Ave =    -4.48  Std =    20.37
   Min TM value:-240 at an offset of 125.725 usec -> 2021-08-22T11:19:34.864.444.450 (XO: 28.864447703)
   Max TM value:208 at an offset of 125.425 usec -> 2021-08-22T11:19:34.864.444.150 (XO: 28.864447403)
   Max dTM/dt [1/us]:1.85e+03 at an offset of 123.350 usec -> 2021-08-22T11:19:34.864.442.075 (XO: 28.864445328)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 125.538 usec -> 2021-08-22T11:19:34.864.444.262 (XO: 28.864447515)
