-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec 22 11:04:41 2021 UTC
-- File 20210822_104110_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.441678 to 23.999354
-each separately from 5s to 23s and from 441678us to 999354us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.557098 in 1 packets: 18
MIU packet time from 2021-08-22T10:41:18.441.678.000 to 2021-08-22T10:41:36.999.354.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.557098 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.4350040 to 24.4349450
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999410 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210822_104110_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.8766825 to 6.8769715
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.5570390  -- OK for SL --
   MIU time from 2021-08-22T10:41:18.441.678.000 to 2021-08-22T10:41:18.441.967.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.557098

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.4340105 to 25.4342995
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-22T10:41:36.999.065.000 to 2021-08-22T10:41:36.999.354.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 6.876682500 corrected by 40bit 10MHz counter of 6.435004000 obtained at 1Hz counter of 5. Result: 5.441679958
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-08-22T10:41:18.441.537.725
   Max 1st sample MIU time among all SL packets: 2021-08-22T10:41:36.999.213.725
   Min 1st sample XO time among all SL packets: 5.441539682
   Max 1st sample XO time among all SL packets: 23.999217422
   TM values from -255 to 217 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-22T10:41:18.441.537.725 (XO: 5.441539682 - 5.441845683, t=37.2 degC)  Rank: 00079
   TM values: Max =   208  Min =  -206  Ave =    -4.52  Std =    21.62
   Min TM value:-206 at an offset of 213.126 usec -> 2021-08-22T10:41:18.441.750.850 (XO: 5.441752808)
   Max TM value:208 at an offset of 111.363 usec -> 2021-08-22T10:41:18.441.649.087 (XO: 5.441651045)
   Max dTM/dt [1/us]:1.52e+03 at an offset of 111.225 usec -> 2021-08-22T10:41:18.441.648.950 (XO: 5.441650907)
   Min dTM/dt [1/us]:-1.49e+03 at an offset of 226.513 usec -> 2021-08-22T10:41:18.441.764.238 (XO: 5.441766195)

 - Block number: 2 starting at 2021-08-22T10:41:36.998.924.725 (XO: 23.998928422 - 23.999234423, t=37.2 degC)  Rank: 00084
   TM values: Max =   217  Min =  -255  Ave =    -4.54  Std =    25.78
   Min TM value:-255 at an offset of 184.988 usec -> 2021-08-22T10:41:36.999.109.713 (XO: 23.999113410)
   Max TM value:217 at an offset of 111.413 usec -> 2021-08-22T10:41:36.999.036.137 (XO: 23.999039834)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 182.738 usec -> 2021-08-22T10:41:36.999.107.463 (XO: 23.999111160)
   Min dTM/dt [1/us]:-2.15e+03 at an offset of 184.838 usec -> 2021-08-22T10:41:36.999.109.563 (XO: 23.999113260)
