-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec 22 08:38:00 2021 UTC
-- File 20210822_083045_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.329679 to 100.626230
-each separately from 6s to 100s and from 210328us to 626230us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.898147 in 1 packets: 18
      19.236974 in 1 packets: 36
      20.936804 in 1 packets: 54
      14.807568 in 1 packets: 72
      12.267879 in 1 packets: 90
      16.147156 in 1 packets: 108
MIU packet time from 2021-08-22T08:30:54.329.679.000 to 2021-08-22T08:32:28.626.230.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.898147 in 1 packets: 18
      19.236974 in 1 packets: 36
      20.936804 in 1 packets: 54
      14.807568 in 1 packets: 72
      12.267879 in 1 packets: 90
      16.147156 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8941522 to 101.8938431
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999639 in 1 packets: 18
      18.9999375 in 1 packets: 36
      20.9999310 in 1 packets: 54
      14.9999506 in 1 packets: 72
      11.9999604 in 1 packets: 90
      15.9999475 in 1 packets: 108
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 90 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
      05 -> 48.5 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Temperature XO: 
      C1 -> 37.2 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 72 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 35 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108, 109, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 72 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 35 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108, 109, ...
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20210822_083045_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.2238324 to 8.2241214
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.8981110  -- OK for SL --
   MIU time from 2021-08-22T08:30:54.329.679.000 to 2021-08-22T08:30:54.329.968.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.898147

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.1222324 to 19.1225214
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.2369110  -- OK for SL --
   MIU time from 2021-08-22T08:31:05.228.115.000 to 2021-08-22T08:31:05.228.404.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.236974

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 38.3594324 to 38.3597214
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.9367350  -- OK for SL --
   MIU time from 2021-08-22T08:31:24.465.378.000 to 2021-08-22T08:31:24.465.667.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.936804

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 59.2964564 to 59.2967454
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.8075190  -- OK for SL --
   MIU time from 2021-08-22T08:31:45.402.471.000 to 2021-08-22T08:31:45.402.760.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.807568

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 74.1042644 to 74.1045534
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.2678390  -- OK for SL --
   MIU time from 2021-08-22T08:32:00.210.328.000 to 2021-08-22T08:32:00.210.617.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.267879

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 86.3723924 to 86.3726814
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.1471030  -- OK for SL --
   MIU time from 2021-08-22T08:32:12.478.496.000 to 2021-08-22T08:32:12.478.785.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.147156

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 102.5197844 to 102.5200734
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-22T08:32:28.625.941.000 to 2021-08-22T08:32:28.626.230.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.223832400 corrected by 40bit 10MHz counter of 7.894152200 obtained at 1Hz counter of 6. Result: 6.329681255
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-08-22T08:30:54.329.538.725
   Max 1st sample MIU time among all SL packets: 2021-08-22T08:32:28.626.089.725
   Min 1st sample XO time among all SL packets: 6.329540980
   Max 1st sample XO time among all SL packets: 100.626092091
   TM values from -248 to 196 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-22T08:30:54.329.538.725 (XO: 6.329540980 - 6.329846981, t=37.2 degC)  Rank: 00077
   TM values: Max =   164  Min =  -235  Ave =    -4.52  Std =    18.17
   Min TM value:-235 at an offset of 128.450 usec -> 2021-08-22T08:30:54.329.667.175 (XO: 6.329669430)
   Max TM value:164 at an offset of 131.063 usec -> 2021-08-22T08:30:54.329.669.787 (XO: 6.329672042)
   Max dTM/dt [1/us]:1.22e+03 at an offset of 130.950 usec -> 2021-08-22T08:30:54.329.669.675 (XO: 6.329671930)
   Min dTM/dt [1/us]:-1.65e+03 at an offset of 115.638 usec -> 2021-08-22T08:30:54.329.654.362 (XO: 6.329656617)

 - Block number: 2 starting at 2021-08-22T08:31:05.227.974.725 (XO: 17.227976777 - 17.228282778, t=37.2 degC)  Rank: 00085
   TM values: Max =   169  Min =  -236  Ave =    -4.34  Std =    19.62
   Min TM value:-236 at an offset of 126.625 usec -> 2021-08-22T08:31:05.228.101.350 (XO: 17.228103403)
   Max TM value:169 at an offset of 129.263 usec -> 2021-08-22T08:31:05.228.103.987 (XO: 17.228106040)
   Max dTM/dt [1/us]:1.17e+03 at an offset of 129.100 usec -> 2021-08-22T08:31:05.228.103.825 (XO: 17.228105878)
   Min dTM/dt [1/us]:-1.54e+03 at an offset of 126.388 usec -> 2021-08-22T08:31:05.228.101.112 (XO: 17.228103165)

 - Block number: 3 starting at 2021-08-22T08:31:24.465.237.725 (XO: 36.465240107 - 36.465546108, t=37.2 degC)  Rank: 00076
   TM values: Max =   181  Min =  -235  Ave =    -4.42  Std =    24.14
   Min TM value:-235 at an offset of 174.588 usec -> 2021-08-22T08:31:24.465.412.313 (XO: 36.465414695)
   Max TM value:181 at an offset of 117.125 usec -> 2021-08-22T08:31:24.465.354.850 (XO: 36.465357232)
   Max dTM/dt [1/us]:1.52e+03 at an offset of 116.938 usec -> 2021-08-22T08:31:24.465.354.662 (XO: 36.465357045)
   Min dTM/dt [1/us]:-1.62e+03 at an offset of 114.300 usec -> 2021-08-22T08:31:24.465.352.025 (XO: 36.465354407)

 - Block number: 4 starting at 2021-08-22T08:31:45.402.330.725 (XO: 57.402332893 - 57.402638894, t=37.2 degC)  Rank: 00076
   TM values: Max =   183  Min =  -227  Ave =    -4.54  Std =    24.44
   Min TM value:-227 at an offset of 126.113 usec -> 2021-08-22T08:31:45.402.456.837 (XO: 57.402459006)
   Max TM value:183 at an offset of 128.688 usec -> 2021-08-22T08:31:45.402.459.412 (XO: 57.402461581)
   Max dTM/dt [1/us]:1.86e+03 at an offset of 128.500 usec -> 2021-08-22T08:31:45.402.459.225 (XO: 57.402461393)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 125.900 usec -> 2021-08-22T08:31:45.402.456.625 (XO: 57.402458793)

 - Block number: 5 starting at 2021-08-22T08:32:00.210.187.725 (XO: 72.210189619 - 72.210495620, t=37.2 degC)  Rank: 00076
   TM values: Max =   185  Min =  -232  Ave =    -4.50  Std =    17.50
   Min TM value:-232 at an offset of 102.075 usec -> 2021-08-22T08:32:00.210.289.800 (XO: 72.210291694)
   Max TM value:185 at an offset of 104.650 usec -> 2021-08-22T08:32:00.210.292.375 (XO: 72.210294269)
   Max dTM/dt [1/us]:1.51e+03 at an offset of 104.500 usec -> 2021-08-22T08:32:00.210.292.225 (XO: 72.210294119)
   Min dTM/dt [1/us]:-1.64e+03 at an offset of 101.925 usec -> 2021-08-22T08:32:00.210.289.650 (XO: 72.210291544)

 - Block number: 6 starting at 2021-08-22T08:32:12.478.355.725 (XO: 84.478358056 - 84.478664057, t=37.2 degC)  Rank: 00087
   TM values: Max =   190  Min =  -248  Ave =    -4.58  Std =    18.41
   Min TM value:-248 at an offset of 115.125 usec -> 2021-08-22T08:32:12.478.470.850 (XO: 84.478473181)
   Max TM value:190 at an offset of 117.638 usec -> 2021-08-22T08:32:12.478.473.362 (XO: 84.478475694)
   Max dTM/dt [1/us]:2.03e+03 at an offset of 117.475 usec -> 2021-08-22T08:32:12.478.473.200 (XO: 84.478475531)
   Min dTM/dt [1/us]:-1.86e+03 at an offset of 114.925 usec -> 2021-08-22T08:32:12.478.470.650 (XO: 84.478472981)

 - Block number: 7 starting at 2021-08-22T08:32:28.625.800.725 (XO: 100.625803090 - 100.626109091, t=37.2 degC)  Rank: 00080
   TM values: Max =   196  Min =  -222  Ave =    -4.45  Std =    27.95
   Min TM value:-222 at an offset of 172.138 usec -> 2021-08-22T08:32:28.625.972.863 (XO: 100.625975228)
   Max TM value:196 at an offset of 114.813 usec -> 2021-08-22T08:32:28.625.915.537 (XO: 100.625917903)
   Max dTM/dt [1/us]:1.47e+03 at an offset of 112.738 usec -> 2021-08-22T08:32:28.625.913.462 (XO: 100.625915828)
   Min dTM/dt [1/us]:-1.78e+03 at an offset of 114.925 usec -> 2021-08-22T08:32:28.625.915.650 (XO: 100.625918016)
