-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Dec 21 22:41:56 2021 UTC
-- File 20210821_103401_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.631238 to 74.991405
-each separately from 0s to 74s and from 271218us to 991405us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.719592 in 1 packets: 18
      19.619727 in 1 packets: 36
      5.299794 in 1 packets: 54
      27.719609 in 1 packets: 72
MIU packet time from 2021-08-21T10:34:04.631.238.000 to 2021-08-21T10:35:18.991.405.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.719592 in 1 packets: 18
      19.619727 in 1 packets: 36
      5.299794 in 1 packets: 54
      27.719609 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.7153986 to 75.7151526
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999269 in 1 packets: 18
      18.9999368 in 1 packets: 36
      5.9999801 in 1 packets: 54
      26.9999102 in 1 packets: 72
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 18 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73
      3 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 18 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73
      3 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210821_103401_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.3466373 to 2.3469263
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.7195190  -- OK for SL --
   MIU time from 2021-08-21T10:34:04.631.238.000 to 2021-08-21T10:34:04.631.527.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.719592

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.0664453 to 24.0667343
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.6196630  -- OK for SL --
   MIU time from 2021-08-21T10:34:26.351.119.000 to 2021-08-21T10:34:26.351.408.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.619727

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.6863973 to 43.6866863
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.2997750  -- OK for SL --
   MIU time from 2021-08-21T10:34:45.971.135.000 to 2021-08-21T10:34:45.971.424.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.299794

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 48.9864613 to 48.9867503
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 27.7195190  -- OK for SL --
   MIU time from 2021-08-21T10:34:51.271.218.000 to 2021-08-21T10:34:51.271.507.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 27.719609

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 76.7062693 to 76.7065583
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-21T10:35:18.991.116.000 to 2021-08-21T10:35:18.991.405.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.346637300 corrected by 40bit 10MHz counter of 1.715398600 obtained at 1Hz counter of 0. Result: 0.631240783
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-21T10:34:04.631.097.725
   Max 1st sample MIU time among all SL packets: 2021-08-21T10:35:18.991.264.725
   Min 1st sample XO time among all SL packets: 0.631100508
   Max 1st sample XO time among all SL packets: 74.991268696
   TM values from -243 to 183 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-21T10:34:04.631.097.725 (XO: 0.631100508 - 0.631406509, t=37.2 degC)  Rank: 00082
   TM values: Max =   168  Min =  -223  Ave =    -4.41  Std =    16.04
   Min TM value:-223 at an offset of 117.275 usec -> 2021-08-21T10:34:04.631.215.000 (XO: 0.631217783)
   Max TM value:168 at an offset of 119.888 usec -> 2021-08-21T10:34:04.631.217.612 (XO: 0.631220396)
   Max dTM/dt [1/us]:1.63e+03 at an offset of 119.675 usec -> 2021-08-21T10:34:04.631.217.400 (XO: 0.631220183)
   Min dTM/dt [1/us]:-1.65e+03 at an offset of 117.038 usec -> 2021-08-21T10:34:04.631.214.762 (XO: 0.631217546)

 - Block number: 2 starting at 2021-08-21T10:34:26.350.978.725 (XO: 22.350980683 - 22.351286684, t=37.2 degC)  Rank: 00079
   TM values: Max =   164  Min =  -225  Ave =    -4.44  Std =    17.24
   Min TM value:-225 at an offset of 117.975 usec -> 2021-08-21T10:34:26.351.096.700 (XO: 22.351098659)
   Max TM value:164 at an offset of 120.600 usec -> 2021-08-21T10:34:26.351.099.325 (XO: 22.351101284)
   Max dTM/dt [1/us]:1.32e+03 at an offset of 120.413 usec -> 2021-08-21T10:34:26.351.099.137 (XO: 22.351101096)
   Min dTM/dt [1/us]:-1.51e+03 at an offset of 117.738 usec -> 2021-08-21T10:34:26.351.096.462 (XO: 22.351098421)

 - Block number: 3 starting at 2021-08-21T10:34:45.970.994.725 (XO: 41.970997929 - 41.971303930, t=37.2 degC)  Rank: 00075
   TM values: Max =   167  Min =  -212  Ave =    -4.44  Std =    21.55
   Min TM value:-212 at an offset of 120.313 usec -> 2021-08-21T10:34:45.971.115.037 (XO: 41.971118242)
   Max TM value:167 at an offset of 120.000 usec -> 2021-08-21T10:34:45.971.114.725 (XO: 41.971117930)
   Max dTM/dt [1/us]:1.54e+03 at an offset of 122.750 usec -> 2021-08-21T10:34:45.971.117.475 (XO: 41.971120680)
   Min dTM/dt [1/us]:-1.71e+03 at an offset of 120.125 usec -> 2021-08-21T10:34:45.971.114.850 (XO: 41.971118055)

 - Block number: 4 starting at 2021-08-21T10:34:51.271.077.725 (XO: 47.271079547 - 47.271385548, t=37.2 degC)  Rank: 00084
   TM values: Max =   183  Min =  -243  Ave =    -4.50  Std =    17.57
   Min TM value:-243 at an offset of 117.063 usec -> 2021-08-21T10:34:51.271.194.787 (XO: 47.271196610)
   Max TM value:183 at an offset of 119.600 usec -> 2021-08-21T10:34:51.271.197.325 (XO: 47.271199147)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 119.425 usec -> 2021-08-21T10:34:51.271.197.150 (XO: 47.271198972)
   Min dTM/dt [1/us]:-1.75e+03 at an offset of 116.863 usec -> 2021-08-21T10:34:51.271.194.587 (XO: 47.271196410)

 - Block number: 5 starting at 2021-08-21T10:35:18.990.975.725 (XO: 74.990979695 - 74.991285696, t=37.2 degC)  Rank: 00087
   TM values: Max =   169  Min =  -222  Ave =    -4.52  Std =    20.53
   Min TM value:-222 at an offset of 118.175 usec -> 2021-08-21T10:35:18.991.093.900 (XO: 74.991097871)
   Max TM value:169 at an offset of 120.838 usec -> 2021-08-21T10:35:18.991.096.562 (XO: 74.991100533)
   Max dTM/dt [1/us]:1.41e+03 at an offset of 120.600 usec -> 2021-08-21T10:35:18.991.096.325 (XO: 74.991100296)
   Min dTM/dt [1/us]:-1.66e+03 at an offset of 105.400 usec -> 2021-08-21T10:35:18.991.081.125 (XO: 74.991085096)
