-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Dec 21 17:57:40 2021 UTC
-- File 20210821_100018_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.295333 to 15.715384
-each separately from 0s to 15s and from 295333us to 715384us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.419473 in 1 packets: 18
MIU packet time from 2021-08-21T10:00:21.295.333.000 to 2021-08-21T10:00:36.715.384.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.419473 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.9429110 to 16.9428612
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999502 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210821_100018_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.2382453 to 2.2385343
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.4194230  -- OK for SL --
   MIU time from 2021-08-21T10:00:21.295.333.000 to 2021-08-21T10:00:21.295.622.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.419473

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.6579573 to 17.6582463
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-21T10:00:36.715.095.000 to 2021-08-21T10:00:36.715.384.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.238245300 corrected by 40bit 10MHz counter of 1.942911000 obtained at 1Hz counter of 0. Result: 0.295335304
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-21T10:00:21.295.192.725
   Max 1st sample MIU time among all SL packets: 2021-08-21T10:00:36.715.243.725
   Min 1st sample XO time among all SL packets: 0.295195029
   Max 1st sample XO time among all SL packets: 15.715247185
   TM values from -253 to 209 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-21T10:00:21.295.192.725 (XO: 0.295195029 - 0.295501030, t=37.2 degC)  Rank: 00077
   TM values: Max =   209  Min =  -253  Ave =    -4.55  Std =    24.98
   Min TM value:-253 at an offset of 128.800 usec -> 2021-08-21T10:00:21.295.321.525 (XO: 0.295323829)
   Max TM value:209 at an offset of 128.488 usec -> 2021-08-21T10:00:21.295.321.212 (XO: 0.295323517)
   Max dTM/dt [1/us]:1.62e+03 at an offset of 131.188 usec -> 2021-08-21T10:00:21.295.323.912 (XO: 0.295326217)
   Min dTM/dt [1/us]:-1.93e+03 at an offset of 128.613 usec -> 2021-08-21T10:00:21.295.321.337 (XO: 0.295323642)

 - Block number: 2 starting at 2021-08-21T10:00:36.714.954.725 (XO: 15.714958184 - 15.715264185, t=37.2 degC)  Rank: 00075
   TM values: Max =   169  Min =  -216  Ave =    -4.47  Std =    23.89
   Min TM value:-216 at an offset of 125.950 usec -> 2021-08-21T10:00:36.715.080.675 (XO: 15.715084135)
   Max TM value:169 at an offset of 232.076 usec -> 2021-08-21T10:00:36.715.186.800 (XO: 15.715190260)
   Max dTM/dt [1/us]:1.52e+03 at an offset of 128.400 usec -> 2021-08-21T10:00:36.715.083.125 (XO: 15.715086585)
   Min dTM/dt [1/us]:-1.59e+03 at an offset of 215.226 usec -> 2021-08-21T10:00:36.715.169.950 (XO: 15.715173410)
