-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Dec 21 08:57:55 2021 UTC
-- File 20210820_114207_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.749423 to 35.349044
-each separately from 0s to 35s and from 348755us to 749712us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.989135 in 1 packets: 18
      10.609619 in 1 packets: 36
MIU packet time from 2021-08-20T11:42:10.749.423.000 to 2021-08-20T11:42:45.349.044.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.989135 in 1 packets: 18
      10.609619 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.9271341 to 36.9270175
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.9999201 in 1 packets: 18
      10.9999633 in 1 packets: 36
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      C0 -> 35.6 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210820_114207_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.6765572 to 2.6768462
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.9890550  -- OK for SL --
   MIU time from 2021-08-20T11:42:10.749.423.000 to 2021-08-20T11:42:10.749.712.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.989135

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.6659012 to 26.6661902
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.6095830  -- OK for SL --
   MIU time from 2021-08-20T11:42:34.738.847.000 to 2021-08-20T11:42:34.739.136.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.609619

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.2757732 to 37.2760622
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-20T11:42:45.348.755.000 to 2021-08-20T11:42:45.349.044.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.676557200 corrected by 40bit 10MHz counter of 1.927134100 obtained at 1Hz counter of 0. Result: 0.749425573
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-20T11:42:10.749.282.725
   Max 1st sample MIU time among all SL packets: 2021-08-20T11:42:45.348.903.725
   Min 1st sample XO time among all SL packets: 0.749285298
   Max 1st sample XO time among all SL packets: 35.348905576
   TM values from -232 to 192 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-20T11:42:10.749.282.725 (XO: 0.749285298 - 0.749591299, t=37.2 degC)  Rank: 00085
   TM values: Max =   185  Min =  -213  Ave =    -4.51  Std =    18.18
   Min TM value:-213 at an offset of 125.850 usec -> 2021-08-20T11:42:10.749.408.575 (XO: 0.749411148)
   Max TM value:185 at an offset of 128.525 usec -> 2021-08-20T11:42:10.749.411.250 (XO: 0.749413823)
   Max dTM/dt [1/us]:1.48e+03 at an offset of 128.388 usec -> 2021-08-20T11:42:10.749.411.112 (XO: 0.749413686)
   Min dTM/dt [1/us]:-1.56e+03 at an offset of 125.613 usec -> 2021-08-20T11:42:10.749.408.337 (XO: 0.749410911)

 - Block number: 2 starting at 2021-08-20T11:42:34.738.706.725 (XO: 24.738709237 - 24.739015238, t=37.2 degC)  Rank: 00075
   TM values: Max =   192  Min =  -227  Ave =    -4.48  Std =    18.43
   Min TM value:-227 at an offset of 127.850 usec -> 2021-08-20T11:42:34.738.834.575 (XO: 24.738837087)
   Max TM value:192 at an offset of 130.463 usec -> 2021-08-20T11:42:34.738.837.187 (XO: 24.738839700)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 125.413 usec -> 2021-08-20T11:42:34.738.832.137 (XO: 24.738834650)
   Min dTM/dt [1/us]:-1.72e+03 at an offset of 127.688 usec -> 2021-08-20T11:42:34.738.834.412 (XO: 24.738836925)

 - Block number: 3 starting at 2021-08-20T11:42:45.348.614.725 (XO: 35.348616575 - 35.348922576, t=35.6 degC)  Rank: 00085
   TM values: Max =   183  Min =  -232  Ave =    -4.50  Std =    18.76
   Min TM value:-232 at an offset of 89.813 usec -> 2021-08-20T11:42:45.348.704.537 (XO: 35.348706388)
   Max TM value:183 at an offset of 92.438 usec -> 2021-08-20T11:42:45.348.707.162 (XO: 35.348709013)
   Max dTM/dt [1/us]:1.61e+03 at an offset of 92.263 usec -> 2021-08-20T11:42:45.348.706.987 (XO: 35.348708838)
   Min dTM/dt [1/us]:-1.85e+03 at an offset of 89.625 usec -> 2021-08-20T11:42:45.348.704.350 (XO: 35.348706201)
