-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Dec 21 06:55:11 2021 UTC
-- File 20210820_104043_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.774468 to 29.602506
-each separately from 8s to 29s and from 602217us to 774757us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.827460 in 1 packets: 18
MIU packet time from 2021-08-20T10:40:54.774.468.000 to 2021-08-20T10:41:15.602.506.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.827460 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.8498138 to 30.8497443
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999305 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210820_104043_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.6242822 to 10.6245712
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.8273910  -- OK for SL --
   MIU time from 2021-08-20T10:40:54.774.468.000 to 2021-08-20T10:40:54.774.757.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.827460

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4519622 to 31.4522512
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-20T10:41:15.602.217.000 to 2021-08-20T10:41:15.602.506.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.624282200 corrected by 40bit 10MHz counter of 9.849813800 obtained at 1Hz counter of 8. Result: 8.774470956
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-20T10:40:54.774.327.725
   Max 1st sample MIU time among all SL packets: 2021-08-20T10:41:15.602.365.725
   Min 1st sample XO time among all SL packets: 8.774330680
   Max 1st sample XO time among all SL packets: 29.602368673
   TM values from -238 to 185 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-20T10:40:54.774.327.725 (XO: 8.774330680 - 8.774636681, t=37.2 degC)  Rank: 00080
   TM values: Max =   185  Min =  -220  Ave =    -4.48  Std =    15.30
   Min TM value:-220 at an offset of 111.850 usec -> 2021-08-20T10:40:54.774.439.575 (XO: 8.774442531)
   Max TM value:185 at an offset of 114.450 usec -> 2021-08-20T10:40:54.774.442.175 (XO: 8.774445131)
   Max dTM/dt [1/us]:1.54e+03 at an offset of 114.275 usec -> 2021-08-20T10:40:54.774.442.000 (XO: 8.774444956)
   Min dTM/dt [1/us]:-1.71e+03 at an offset of 111.700 usec -> 2021-08-20T10:40:54.774.439.425 (XO: 8.774442381)

 - Block number: 2 starting at 2021-08-20T10:41:15.602.076.725 (XO: 29.602079672 - 29.602385673, t=37.2 degC)  Rank: 00083
   TM values: Max =   167  Min =  -238  Ave =    -4.40  Std =    18.38
   Min TM value:-238 at an offset of 122.450 usec -> 2021-08-20T10:41:15.602.199.175 (XO: 29.602202122)
   Max TM value:167 at an offset of 125.100 usec -> 2021-08-20T10:41:15.602.201.825 (XO: 29.602204772)
   Max dTM/dt [1/us]:1.62e+03 at an offset of 124.875 usec -> 2021-08-20T10:41:15.602.201.600 (XO: 29.602204547)
   Min dTM/dt [1/us]:-1.68e+03 at an offset of 122.250 usec -> 2021-08-20T10:41:15.602.198.975 (XO: 29.602201922)
