-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Dec 21 03:14:02 2021 UTC
-- File 20210819_164710_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.062126 to 46.451809
-each separately from 2s to 46s and from 62126us to 769014us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.414117 in 1 packets: 18
      10.291904 in 1 packets: 36
      12.682506 in 1 packets: 54
MIU packet time from 2021-08-19T16:47:15.062.126.000 to 2021-08-19T16:47:59.451.809.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.414117 in 1 packets: 18
      10.291904 in 1 packets: 36
      12.682506 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.9353549 to 47.9352085
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999301 in 1 packets: 18
      9.9999667 in 1 packets: 36
      12.9999568 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 72 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210819_164710_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.9974817 to 3.9977707
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.4140470  -- OK for SL --
   MIU time from 2021-08-19T16:47:15.062.126.000 to 2021-08-19T16:47:15.062.415.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.414117

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.4118177 to 25.4121067
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.2918710  -- OK for SL --
   MIU time from 2021-08-19T16:47:36.476.532.000 to 2021-08-19T16:47:36.476.821.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.291904

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.7039777 to 35.7042667
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.6824630  -- OK for SL --
   MIU time from 2021-08-19T16:47:46.768.725.000 to 2021-08-19T16:47:46.769.014.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.682506

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 48.3867297 to 48.3870187
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-19T16:47:59.451.520.000 to 2021-08-19T16:47:59.451.809.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 3.997481700 corrected by 40bit 10MHz counter of 3.935354900 obtained at 1Hz counter of 2. Result: 2.062127005
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-19T16:47:15.061.985.725
   Max 1st sample MIU time among all SL packets: 2021-08-19T16:47:59.451.668.725
   Min 1st sample XO time among all SL packets: 2.061986730
   Max 1st sample XO time among all SL packets: 46.451671461
   TM values from -256 to 244 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-19T16:47:15.061.985.725 (XO: 2.061986730 - 2.062292731, t=37.2 degC)  Rank: 00075
   TM values: Max =   207  Min =  -249  Ave =    -4.49  Std =    20.15
   Min TM value:-249 at an offset of 125.738 usec -> 2021-08-19T16:47:15.062.111.462 (XO: 2.062112467)
   Max TM value:207 at an offset of 125.438 usec -> 2021-08-19T16:47:15.062.111.162 (XO: 2.062112167)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 128.088 usec -> 2021-08-19T16:47:15.062.113.812 (XO: 2.062114817)
   Min dTM/dt [1/us]:-2.01e+03 at an offset of 125.538 usec -> 2021-08-19T16:47:15.062.111.262 (XO: 2.062112267)

 - Block number: 2 starting at 2021-08-19T16:47:36.476.391.725 (XO: 23.476393997 - 23.476699998, t=37.2 degC)  Rank: 00081
   TM values: Max =   244  Min =  -223  Ave =    -4.56  Std =    20.79
   Min TM value:-223 at an offset of 104.225 usec -> 2021-08-19T16:47:36.476.495.950 (XO: 23.476498222)
   Max TM value:244 at an offset of 106.725 usec -> 2021-08-19T16:47:36.476.498.450 (XO: 23.476500722)
   Max dTM/dt [1/us]:2.05e+03 at an offset of 101.925 usec -> 2021-08-19T16:47:36.476.493.650 (XO: 23.476495922)
   Min dTM/dt [1/us]:-2.19e+03 at an offset of 104.075 usec -> 2021-08-19T16:47:36.476.495.800 (XO: 23.476498072)

 - Block number: 3 starting at 2021-08-19T16:47:46.768.584.725 (XO: 33.768588261 - 33.768894262, t=37.2 degC)  Rank: 00078
   TM values: Max =   192  Min =  -256  Ave =    -4.51  Std =    20.46
   Min TM value:-256 at an offset of 127.888 usec -> 2021-08-19T16:47:46.768.712.612 (XO: 33.768716149)
   Max TM value:192 at an offset of 127.588 usec -> 2021-08-19T16:47:46.768.712.312 (XO: 33.768715849)
   Max dTM/dt [1/us]:1.85e+03 at an offset of 130.250 usec -> 2021-08-19T16:47:46.768.714.975 (XO: 33.768718512)
   Min dTM/dt [1/us]:-1.97e+03 at an offset of 127.675 usec -> 2021-08-19T16:47:46.768.712.400 (XO: 33.768715937)

 - Block number: 4 starting at 2021-08-19T16:47:59.451.379.725 (XO: 46.451382460 - 46.451688461, t=37.2 degC)  Rank: 00082
   TM values: Max =   206  Min =  -238  Ave =    -4.54  Std =    20.84
   Min TM value:-238 at an offset of 111.900 usec -> 2021-08-19T16:47:59.451.491.625 (XO: 46.451494360)
   Max TM value:206 at an offset of 127.063 usec -> 2021-08-19T16:47:59.451.506.787 (XO: 46.451509523)
   Max dTM/dt [1/us]:1.53e+03 at an offset of 126.888 usec -> 2021-08-19T16:47:59.451.506.612 (XO: 46.451509348)
   Min dTM/dt [1/us]:-1.85e+03 at an offset of 124.288 usec -> 2021-08-19T16:47:59.451.504.012 (XO: 46.451506748)
