-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Dec 21 02:22:10 2021 UTC
-- File 20210819_161815_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.245035 to 55.044511
-each separately from 11s to 55s and from 44222us to 627387us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.381774 in 1 packets: 18
      22.840779 in 1 packets: 36
      11.575767 in 1 packets: 54
MIU packet time from 2021-08-19T16:18:29.245.035.000 to 2021-08-19T16:19:13.044.511.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.381774 in 1 packets: 18
      22.840779 in 1 packets: 36
      11.575767 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.9030138 to 56.9028673
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999699 in 1 packets: 18
      22.9999234 in 1 packets: 36
      11.9999602 in 1 packets: 54
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 71 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 71 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210819_161815_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.1480503 to 13.1483393
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.3817430  -- OK for SL --
   MIU time from 2021-08-19T16:18:29.245.035.000 to 2021-08-19T16:18:29.245.324.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.381774

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.5300823 to 22.5303713
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.8407030  -- OK for SL --
   MIU time from 2021-08-19T16:18:38.627.098.000 to 2021-08-19T16:18:38.627.387.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.840779

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.3710743 to 45.3713633
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.5757270  -- OK for SL --
   MIU time from 2021-08-19T16:19:01.468.166.000 to 2021-08-19T16:19:01.468.455.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.575767

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 56.9470903 to 56.9473793
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-19T16:19:13.044.222.000 to 2021-08-19T16:19:13.044.511.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.148050300 corrected by 40bit 10MHz counter of 12.903013800 obtained at 1Hz counter of 11. Result: 11.245037309
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-19T16:18:29.244.894.725
   Max 1st sample MIU time among all SL packets: 2021-08-19T16:19:13.044.370.725
   Min 1st sample XO time among all SL packets: 11.244897033
   Max 1st sample XO time among all SL packets: 55.044371871
   TM values from -254 to 252 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-19T16:18:29.244.894.725 (XO: 11.244897033 - 11.245203034, t=37.2 degC)  Rank: 00083
   TM values: Max =   208  Min =  -236  Ave =    -4.56  Std =    25.80
   Min TM value:-236 at an offset of 121.463 usec -> 2021-08-19T16:18:29.245.016.187 (XO: 11.245018496)
   Max TM value:208 at an offset of 124.113 usec -> 2021-08-19T16:18:29.245.018.837 (XO: 11.245021146)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 123.925 usec -> 2021-08-19T16:18:29.245.018.650 (XO: 11.245020959)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 121.263 usec -> 2021-08-19T16:18:29.245.015.987 (XO: 11.245018296)

 - Block number: 2 starting at 2021-08-19T16:18:38.626.957.725 (XO: 20.626960394 - 20.627266395, t=37.2 degC)  Rank: 00074
   TM values: Max =   203  Min =  -249  Ave =    -4.61  Std =    19.61
   Min TM value:-249 at an offset of 111.338 usec -> 2021-08-19T16:18:38.627.069.062 (XO: 20.627071732)
   Max TM value:203 at an offset of 113.988 usec -> 2021-08-19T16:18:38.627.071.712 (XO: 20.627074382)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 113.750 usec -> 2021-08-19T16:18:38.627.071.475 (XO: 20.627074144)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 111.150 usec -> 2021-08-19T16:18:38.627.068.875 (XO: 20.627071544)

 - Block number: 3 starting at 2021-08-19T16:19:01.468.025.725 (XO: 43.468028469 - 43.468334471, t=37.2 degC)  Rank: 00078
   TM values: Max =   252  Min =  -254  Ave =    -4.49  Std =    27.97
   Min TM value:-254 at an offset of 126.413 usec -> 2021-08-19T16:19:01.468.152.137 (XO: 43.468154882)
   Max TM value:252 at an offset of 166.763 usec -> 2021-08-19T16:19:01.468.192.488 (XO: 43.468195233)
   Max dTM/dt [1/us]:2.00e+03 at an offset of 166.638 usec -> 2021-08-19T16:19:01.468.192.363 (XO: 43.468195108)
   Min dTM/dt [1/us]:-1.92e+03 at an offset of 181.938 usec -> 2021-08-19T16:19:01.468.207.663 (XO: 43.468210408)

 - Block number: 4 starting at 2021-08-19T16:19:13.044.081.725 (XO: 55.044082870 - 55.044388871, t=37.2 degC)  Rank: 00079
   TM values: Max =   201  Min =  -241  Ave =    -4.56  Std =    25.50
   Min TM value:-241 at an offset of 108.388 usec -> 2021-08-19T16:19:13.044.190.112 (XO: 55.044191258)
   Max TM value:201 at an offset of 33.325 usec -> 2021-08-19T16:19:13.044.115.050 (XO: 55.044116196)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 123.375 usec -> 2021-08-19T16:19:13.044.205.100 (XO: 55.044206246)
   Min dTM/dt [1/us]:-1.81e+03 at an offset of 33.438 usec -> 2021-08-19T16:19:13.044.115.162 (XO: 55.044116308)
