-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Dec 20 22:35:03 2021 UTC
-- File 20210819_124207_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.859456 to 17.186738
-each separately from 2s to 17s and from 186449us to 859745us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.326704 in 1 packets: 18
MIU packet time from 2021-08-19T12:42:12.859.456.000 to 2021-08-19T12:42:27.186.738.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.326704 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.9707724 to 18.9707229
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999505 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210819_124207_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.8302293 to 4.8305183
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.3266550  -- OK for SL --
   MIU time from 2021-08-19T12:42:12.859.456.000 to 2021-08-19T12:42:12.859.745.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.326704

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.1571733 to 19.1574623
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-19T12:42:27.186.449.000 to 2021-08-19T12:42:27.186.738.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.830229300 corrected by 40bit 10MHz counter of 3.970772400 obtained at 1Hz counter of 2. Result: 2.859459736
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-19T12:42:12.859.315.725
   Max 1st sample MIU time among all SL packets: 2021-08-19T12:42:27.186.597.725
   Min 1st sample XO time among all SL packets: 2.859319461
   Max 1st sample XO time among all SL packets: 17.186599741
   TM values from -245 to 209 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-19T12:42:12.859.315.725 (XO: 2.859319461 - 2.859625462, t=35.6 degC)  Rank: 00081
   TM values: Max =   209  Min =  -231  Ave =    -4.49  Std =    19.69
   Min TM value:-231 at an offset of 105.163 usec -> 2021-08-19T12:42:12.859.420.887 (XO: 2.859424624)
   Max TM value:209 at an offset of 107.825 usec -> 2021-08-19T12:42:12.859.423.550 (XO: 2.859427286)
   Max dTM/dt [1/us]:1.74e+03 at an offset of 107.650 usec -> 2021-08-19T12:42:12.859.423.375 (XO: 2.859427111)
   Min dTM/dt [1/us]:-1.97e+03 at an offset of 104.950 usec -> 2021-08-19T12:42:12.859.420.675 (XO: 2.859424411)

 - Block number: 2 starting at 2021-08-19T12:42:27.186.308.725 (XO: 17.186310740 - 17.186616741, t=35.6 degC)  Rank: 00078
   TM values: Max =   196  Min =  -245  Ave =    -4.56  Std =    18.44
   Min TM value:-245 at an offset of 120.950 usec -> 2021-08-19T12:42:27.186.429.675 (XO: 17.186431690)
   Max TM value:196 at an offset of 123.550 usec -> 2021-08-19T12:42:27.186.432.275 (XO: 17.186434290)
   Max dTM/dt [1/us]:1.56e+03 at an offset of 123.388 usec -> 2021-08-19T12:42:27.186.432.112 (XO: 17.186434128)
   Min dTM/dt [1/us]:-1.87e+03 at an offset of 120.788 usec -> 2021-08-19T12:42:27.186.429.512 (XO: 17.186431528)
