-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Dec 20 16:12:03 2021 UTC
-- File 20210819_115156_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.457594 to 17.627378
-each separately from 10s to 17s and from 457594us to 627378us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.169206 in 1 packets: 18
MIU packet time from 2021-08-19T11:52:09.457.594.000 to 2021-08-19T11:52:16.627.378.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.169206 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.9374653 to 18.9374423
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.9999770 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210819_115156_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.3950598 to 12.3953488
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.1691830  -- OK for SL --
   MIU time from 2021-08-19T11:52:09.457.594.000 to 2021-08-19T11:52:09.457.883.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.169206

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.5645318 to 19.5648208
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-19T11:52:16.627.089.000 to 2021-08-19T11:52:16.627.378.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.395059800 corrected by 40bit 10MHz counter of 11.937465300 obtained at 1Hz counter of 10. Result: 10.457596010
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-19T11:52:09.457.453.725
   Max 1st sample MIU time among all SL packets: 2021-08-19T11:52:16.627.237.725
   Min 1st sample XO time among all SL packets: 10.457455735
   Max 1st sample XO time among all SL packets: 17.627240358
   TM values from -238 to 205 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-19T11:52:09.457.453.725 (XO: 10.457455735 - 10.457761736, t=37.2 degC)  Rank: 00084
   TM values: Max =   178  Min =  -238  Ave =    -4.49  Std =    22.32
   Min TM value:-238 at an offset of 125.925 usec -> 2021-08-19T11:52:09.457.579.650 (XO: 10.457581660)
   Max TM value:178 at an offset of 169.401 usec -> 2021-08-19T11:52:09.457.623.125 (XO: 10.457625135)
   Max dTM/dt [1/us]:1.50e+03 at an offset of 128.363 usec -> 2021-08-19T11:52:09.457.582.087 (XO: 10.457584098)
   Min dTM/dt [1/us]:-1.91e+03 at an offset of 169.526 usec -> 2021-08-19T11:52:09.457.623.250 (XO: 10.457625260)

 - Block number: 2 starting at 2021-08-19T11:52:16.626.948.725 (XO: 17.626951357 - 17.627257358, t=37.2 degC)  Rank: 00080
   TM values: Max =   205  Min =  -233  Ave =    -4.47  Std =    19.21
   Min TM value:-233 at an offset of 125.600 usec -> 2021-08-19T11:52:16.627.074.325 (XO: 17.627076957)
   Max TM value:205 at an offset of 125.288 usec -> 2021-08-19T11:52:16.627.074.012 (XO: 17.627076645)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 127.950 usec -> 2021-08-19T11:52:16.627.076.675 (XO: 17.627079307)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 125.388 usec -> 2021-08-19T11:52:16.627.074.112 (XO: 17.627076745)
