-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Dec 20 08:43:27 2021 UTC
-- File 20210818_152626_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.184523 to 15.204816
-each separately from 14s to 15s and from 184523us to 204816us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      1.019715 in 1 packets: 18
MIU packet time from 2021-08-18T15:26:43.184.523.000 to 2021-08-18T15:26:44.204.816.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      1.019715 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.9308255 to 16.9308222
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      0.9999967 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210818_152626_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.1153499 to 16.1156389
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 1.0197110  -- OK for SL --
   MIU time from 2021-08-18T15:26:43.184.523.000 to 2021-08-18T15:26:43.184.812.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 1.019715

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.1353499 to 17.1356389
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-18T15:26:44.204.527.000 to 2021-08-18T15:26:44.204.816.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.115349900 corrected by 40bit 10MHz counter of 15.930825500 obtained at 1Hz counter of 14. Result: 14.184525009
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-18T15:26:43.184.382.725
   Max 1st sample MIU time among all SL packets: 2021-08-18T15:26:44.204.675.725
   Min 1st sample XO time among all SL packets: 14.184384733
   Max 1st sample XO time among all SL packets: 15.204677100
   TM values from -265 to 213 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-18T15:26:43.184.382.725 (XO: 14.184384733 - 14.184690734, t=37.2 degC)  Rank: 00085
   TM values: Max =   211  Min =  -265  Ave =    -4.53  Std =    20.65
   Min TM value:-265 at an offset of 115.275 usec -> 2021-08-18T15:26:43.184.498.000 (XO: 14.184500009)
   Max TM value:211 at an offset of 115.000 usec -> 2021-08-18T15:26:43.184.497.725 (XO: 14.184499734)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 112.875 usec -> 2021-08-18T15:26:43.184.495.600 (XO: 14.184497609)
   Min dTM/dt [1/us]:-1.97e+03 at an offset of 115.125 usec -> 2021-08-18T15:26:43.184.497.850 (XO: 14.184499859)

 - Block number: 2 starting at 2021-08-18T15:26:44.204.386.725 (XO: 15.204388099 - 15.204694100, t=37.2 degC)  Rank: 00082
   TM values: Max =   213  Min =  -253  Ave =    -4.65  Std =    17.76
   Min TM value:-253 at an offset of 114.788 usec -> 2021-08-18T15:26:44.204.501.512 (XO: 15.204502887)
   Max TM value:213 at an offset of 117.425 usec -> 2021-08-18T15:26:44.204.504.150 (XO: 15.204505525)
   Max dTM/dt [1/us]:1.55e+03 at an offset of 117.250 usec -> 2021-08-18T15:26:44.204.503.975 (XO: 15.204505350)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 114.650 usec -> 2021-08-18T15:26:44.204.501.375 (XO: 15.204502750)
