-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Dec 20 01:29:19 2021 UTC
-- File 20210818_025741_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.389977 to 103.228800
-each separately from 5s to 103s and from 9625us to 509371us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.869451 in 1 packets: 18
      16.889767 in 1 packets: 36
      17.859563 in 1 packets: 54
      8.099642 in 1 packets: 72
      21.399237 in 1 packets: 90
      15.719140 in 1 packets: 108
MIU packet time from 2021-08-18T02:57:49.389.977.000 to 2021-08-18T02:59:27.228.800.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.869451 in 1 packets: 18
      16.889767 in 1 packets: 36
      17.859563 in 1 packets: 54
      8.099642 in 1 packets: 72
      21.399237 in 1 packets: 90
      15.719140 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.8843986 to 104.8840732
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999405 in 1 packets: 18
      16.9999436 in 1 packets: 36
      17.9999402 in 1 packets: 54
      7.9999735 in 1 packets: 72
      20.9999300 in 1 packets: 90
      15.9999468 in 1 packets: 108
Temperature FPGA: 
      04 -> 49.8 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
      05 -> 48.5 degC in 18 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Temperature XO: 
      C1 -> 37.2 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
      2 in 53 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, 91, ...
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
      2 in 53 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, 91, ...
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20210818_025741_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.2743770 to 7.2746660
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.8693910  -- OK for SL --
   MIU time from 2021-08-18T02:57:49.389.977.000 to 2021-08-18T02:57:49.390.266.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.869451

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.1440570 to 25.1443460
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.8897110  -- OK for SL --
   MIU time from 2021-08-18T02:58:07.259.717.000 to 2021-08-18T02:58:07.260.006.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.889767

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.0340570 to 42.0343460
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.8595030  -- OK for SL --
   MIU time from 2021-08-18T02:58:24.149.773.000 to 2021-08-18T02:58:24.150.062.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.859563

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 59.8938490 to 59.8941380
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.0996150  -- OK for SL --
   MIU time from 2021-08-18T02:58:42.009.625.000 to 2021-08-18T02:58:42.009.914.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.099642

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 67.9937530 to 67.9940420
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.3991670  -- OK for SL --
   MIU time from 2021-08-18T02:58:50.109.556.000 to 2021-08-18T02:58:50.109.845.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.399237

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 89.3932090 to 89.3934980
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.7190870  -- OK for SL --
   MIU time from 2021-08-18T02:59:11.509.082.000 to 2021-08-18T02:59:11.509.371.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.719140

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 105.1125850 to 105.1128740
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-18T02:59:27.228.511.000 to 2021-08-18T02:59:27.228.800.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.274377000 corrected by 40bit 10MHz counter of 6.884398600 obtained at 1Hz counter of 5. Result: 5.389979687
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-18T02:57:49.389.836.725
   Max 1st sample MIU time among all SL packets: 2021-08-18T02:59:27.228.659.725
   Min 1st sample XO time among all SL packets: 5.389839411
   Max 1st sample XO time among all SL packets: 103.228661280
   TM values from -261 to 219 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-18T02:57:49.389.836.725 (XO: 5.389839411 - 5.390145412, t=37.2 degC)  Rank: 00080
   TM values: Max =   179  Min =  -226  Ave =    -4.44  Std =    25.94
   Min TM value:-226 at an offset of 118.288 usec -> 2021-08-18T02:57:49.389.955.012 (XO: 5.389957699)
   Max TM value:179 at an offset of 120.938 usec -> 2021-08-18T02:57:49.389.957.662 (XO: 5.389960349)
   Max dTM/dt [1/us]:1.54e+03 at an offset of 120.738 usec -> 2021-08-18T02:57:49.389.957.462 (XO: 5.389960149)
   Min dTM/dt [1/us]:-1.77e+03 at an offset of 192.138 usec -> 2021-08-18T02:57:49.390.028.863 (XO: 5.390031550)

 - Block number: 2 starting at 2021-08-18T02:58:07.259.576.725 (XO: 23.259578482 - 23.259884483, t=37.2 degC)  Rank: 00079
   TM values: Max =   206  Min =  -225  Ave =    -4.55  Std =    20.82
   Min TM value:-225 at an offset of 105.888 usec -> 2021-08-18T02:58:07.259.682.612 (XO: 23.259684369)
   Max TM value:206 at an offset of 103.825 usec -> 2021-08-18T02:58:07.259.680.550 (XO: 23.259682307)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 103.663 usec -> 2021-08-18T02:58:07.259.680.387 (XO: 23.259682144)
   Min dTM/dt [1/us]:-1.85e+03 at an offset of 93.225 usec -> 2021-08-18T02:58:07.259.669.950 (XO: 23.259671707)

 - Block number: 3 starting at 2021-08-18T02:58:24.149.632.725 (XO: 40.149634519 - 40.149940520, t=37.2 degC)  Rank: 00081
   TM values: Max =   181  Min =  -233  Ave =    -4.52  Std =    20.09
   Min TM value:-233 at an offset of 118.700 usec -> 2021-08-18T02:58:24.149.751.425 (XO: 40.149753219)
   Max TM value:181 at an offset of 121.288 usec -> 2021-08-18T02:58:24.149.754.012 (XO: 40.149755807)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 121.100 usec -> 2021-08-18T02:58:24.149.753.825 (XO: 40.149755619)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 118.525 usec -> 2021-08-18T02:58:24.149.751.250 (XO: 40.149753044)

 - Block number: 4 starting at 2021-08-18T02:58:42.009.484.725 (XO: 58.009485856 - 58.009791857, t=37.2 degC)  Rank: 00072
   TM values: Max =   188  Min =  -240  Ave =    -4.44  Std =    19.84
   Min TM value:-240 at an offset of 93.550 usec -> 2021-08-18T02:58:42.009.578.275 (XO: 58.009579407)
   Max TM value:188 at an offset of 96.113 usec -> 2021-08-18T02:58:42.009.580.837 (XO: 58.009581969)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 95.950 usec -> 2021-08-18T02:58:42.009.580.675 (XO: 58.009581807)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 93.388 usec -> 2021-08-18T02:58:42.009.578.112 (XO: 58.009579244)

 - Block number: 5 starting at 2021-08-18T02:58:50.109.415.725 (XO: 66.109416686 - 66.109722687, t=37.2 degC)  Rank: 00082
   TM values: Max =   207  Min =  -229  Ave =    -4.49  Std =    21.46
   Min TM value:-229 at an offset of 107.975 usec -> 2021-08-18T02:58:50.109.523.700 (XO: 66.109524661)
   Max TM value:207 at an offset of 107.700 usec -> 2021-08-18T02:58:50.109.523.425 (XO: 66.109524386)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 110.338 usec -> 2021-08-18T02:58:50.109.526.062 (XO: 66.109527024)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 107.800 usec -> 2021-08-18T02:58:50.109.523.525 (XO: 66.109524486)

 - Block number: 6 starting at 2021-08-18T02:59:11.508.941.725 (XO: 87.508944005 - 87.509250006, t=37.2 degC)  Rank: 00077
   TM values: Max =   177  Min =  -223  Ave =    -4.42  Std =    20.83
   Min TM value:-223 at an offset of 118.438 usec -> 2021-08-18T02:59:11.509.060.162 (XO: 87.509062442)
   Max TM value:177 at an offset of 121.125 usec -> 2021-08-18T02:59:11.509.062.850 (XO: 87.509065130)
   Max dTM/dt [1/us]:1.37e+03 at an offset of 120.863 usec -> 2021-08-18T02:59:11.509.062.587 (XO: 87.509064867)
   Min dTM/dt [1/us]:-1.53e+03 at an offset of 91.338 usec -> 2021-08-18T02:59:11.509.033.062 (XO: 87.509035342)

 - Block number: 7 starting at 2021-08-18T02:59:27.228.370.725 (XO: 103.228372279 - 103.228678280, t=37.2 degC)  Rank: 00080
   TM values: Max =   219  Min =  -261  Ave =    -4.48  Std =    29.26
   Min TM value:-261 at an offset of 120.763 usec -> 2021-08-18T02:59:27.228.491.487 (XO: 103.228493042)
   Max TM value:219 at an offset of 120.438 usec -> 2021-08-18T02:59:27.228.491.162 (XO: 103.228492717)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 147.463 usec -> 2021-08-18T02:59:27.228.518.187 (XO: 103.228519742)
   Min dTM/dt [1/us]:-2.21e+03 at an offset of 120.563 usec -> 2021-08-18T02:59:27.228.491.287 (XO: 103.228492842)
