-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Dec 19 23:40:20 2021 UTC
-- File 20210818_010837_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.349796 to 71.047913
-each separately from 2s to 71s and from 47624us to 934473us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.072036 in 1 packets: 18
      19.511774 in 1 packets: 36
      13.541085 in 1 packets: 54
      19.571777 in 1 packets: 72
MIU packet time from 2021-08-18T01:08:42.349.796.000 to 2021-08-18T01:09:51.047.913.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.072036 in 1 packets: 18
      19.511774 in 1 packets: 36
      13.541085 in 1 packets: 54
      19.571777 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.9350030 to 72.9347752
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999472 in 1 packets: 18
      18.9999372 in 1 packets: 36
      13.9999538 in 1 packets: 54
      19.9999340 in 1 packets: 72
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210818_010837_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.2848001 to 4.2850891
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.0719830  -- OK for SL --
   MIU time from 2021-08-18T01:08:42.349.796.000 to 2021-08-18T01:08:42.350.085.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.072036

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.3570721 to 20.3573611
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.5117110  -- OK for SL --
   MIU time from 2021-08-18T01:08:58.422.121.000 to 2021-08-18T01:08:58.422.410.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.511774

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.8690721 to 39.8693611
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.5410390  -- OK for SL --
   MIU time from 2021-08-18T01:09:17.934.184.000 to 2021-08-18T01:09:17.934.473.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.541085

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 53.4104001 to 53.4106891
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.5717110  -- OK for SL --
   MIU time from 2021-08-18T01:09:31.475.558.000 to 2021-08-18T01:09:31.475.847.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.571777

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 72.9824001 to 72.9826891
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-18T01:09:51.047.624.000 to 2021-08-18T01:09:51.047.913.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.284800100 corrected by 40bit 10MHz counter of 3.935003000 obtained at 1Hz counter of 2. Result: 2.349798254
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-18T01:08:42.349.655.725
   Max 1st sample MIU time among all SL packets: 2021-08-18T01:09:51.047.772.725
   Min 1st sample XO time among all SL packets: 2.349657979
   Max 1st sample XO time among all SL packets: 71.047773783
   TM values from -261 to 219 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-18T01:08:42.349.655.725 (XO: 2.349657979 - 2.349963980, t=37.2 degC)  Rank: 00073
   TM values: Max =   183  Min =  -222  Ave =    -4.46  Std =    22.14
   Min TM value:-222 at an offset of 123.350 usec -> 2021-08-18T01:08:42.349.779.075 (XO: 2.349781329)
   Max TM value:183 at an offset of 126.075 usec -> 2021-08-18T01:08:42.349.781.800 (XO: 2.349784054)
   Max dTM/dt [1/us]:1.62e+03 at an offset of 125.913 usec -> 2021-08-18T01:08:42.349.781.637 (XO: 2.349783892)
   Min dTM/dt [1/us]:-1.49e+03 at an offset of 96.113 usec -> 2021-08-18T01:08:42.349.751.837 (XO: 2.349754092)

 - Block number: 2 starting at 2021-08-18T01:08:58.421.980.725 (XO: 18.421983018 - 18.422289019, t=37.2 degC)  Rank: 00080
   TM values: Max =   209  Min =  -230  Ave =    -4.53  Std =    20.84
   Min TM value:-230 at an offset of 124.863 usec -> 2021-08-18T01:08:58.422.105.587 (XO: 18.422107880)
   Max TM value:209 at an offset of 124.600 usec -> 2021-08-18T01:08:58.422.105.325 (XO: 18.422107618)
   Max dTM/dt [1/us]:1.93e+03 at an offset of 122.513 usec -> 2021-08-18T01:08:58.422.103.237 (XO: 18.422105530)
   Min dTM/dt [1/us]:-1.93e+03 at an offset of 124.700 usec -> 2021-08-18T01:08:58.422.105.425 (XO: 18.422107718)

 - Block number: 3 starting at 2021-08-18T01:09:17.934.043.725 (XO: 37.934047507 - 37.934353508, t=37.2 degC)  Rank: 00081
   TM values: Max =   192  Min =  -226  Ave =    -4.45  Std =    18.45
   Min TM value:-226 at an offset of 123.450 usec -> 2021-08-18T01:09:17.934.167.175 (XO: 37.934170958)
   Max TM value:192 at an offset of 126.125 usec -> 2021-08-18T01:09:17.934.169.850 (XO: 37.934173633)
   Max dTM/dt [1/us]:1.75e+03 at an offset of 125.950 usec -> 2021-08-18T01:09:17.934.169.675 (XO: 37.934173458)
   Min dTM/dt [1/us]:-1.76e+03 at an offset of 123.263 usec -> 2021-08-18T01:09:17.934.166.987 (XO: 37.934170770)

 - Block number: 4 starting at 2021-08-18T01:09:31.475.417.725 (XO: 51.475420194 - 51.475726195, t=37.2 degC)  Rank: 00081
   TM values: Max =   219  Min =  -261  Ave =    -4.55  Std =    29.72
   Min TM value:-261 at an offset of 142.525 usec -> 2021-08-18T01:09:31.475.560.250 (XO: 51.475562719)
   Max TM value:219 at an offset of 140.288 usec -> 2021-08-18T01:09:31.475.558.012 (XO: 51.475560482)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 140.125 usec -> 2021-08-18T01:09:31.475.557.850 (XO: 51.475560319)
   Min dTM/dt [1/us]:-2.36e+03 at an offset of 113.800 usec -> 2021-08-18T01:09:31.475.531.525 (XO: 51.475533994)

 - Block number: 5 starting at 2021-08-18T01:09:51.047.483.725 (XO: 71.047484782 - 71.047790783, t=37.2 degC)  Rank: 00078
   TM values: Max =   166  Min =  -215  Ave =    -4.48  Std =    21.95
   Min TM value:-215 at an offset of 109.175 usec -> 2021-08-18T01:09:51.047.592.900 (XO: 71.047593957)
   Max TM value:166 at an offset of 108.838 usec -> 2021-08-18T01:09:51.047.592.562 (XO: 71.047593620)
   Max dTM/dt [1/us]:1.57e+03 at an offset of 124.938 usec -> 2021-08-18T01:09:51.047.608.662 (XO: 71.047609720)
   Min dTM/dt [1/us]:-1.56e+03 at an offset of 108.988 usec -> 2021-08-18T01:09:51.047.592.712 (XO: 71.047593770)
