-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Dec 18 23:22:08 2021 UTC
-- File 20210816_233516_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.823286 to 17.653103
-each separately from 0s to 17s and from 652814us to 823575us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.829239 in 1 packets: 18
MIU packet time from 2021-08-16T23:35:19.823.286.000 to 2021-08-16T23:35:36.653.103.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.829239 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.6213632 to 18.6213071
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999439 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210816_233516_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.4446503 to 2.4449393
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.8291830  -- OK for SL --
   MIU time from 2021-08-16T23:35:19.823.286.000 to 2021-08-16T23:35:19.823.575.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.829239

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.2741223 to 19.2744113
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-16T23:35:36.652.814.000 to 2021-08-16T23:35:36.653.103.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.444650300 corrected by 40bit 10MHz counter of 1.621363200 obtained at 1Hz counter of 0. Result: 0.823289817
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-16T23:35:19.823.145.725
   Max 1st sample MIU time among all SL packets: 2021-08-16T23:35:36.652.962.725
   Min 1st sample XO time among all SL packets: 0.823149541
   Max 1st sample XO time among all SL packets: 17.652966080
   TM values from -256 to 204 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-16T23:35:19.823.145.725 (XO: 0.823149541 - 0.823455542, t=37.2 degC)  Rank: 00085
   TM values: Max =   195  Min =  -256  Ave =    -4.59  Std =    20.70
   Min TM value:-256 at an offset of 108.638 usec -> 2021-08-16T23:35:19.823.254.362 (XO: 0.823258179)
   Max TM value:195 at an offset of 108.338 usec -> 2021-08-16T23:35:19.823.254.062 (XO: 0.823257879)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 111.025 usec -> 2021-08-16T23:35:19.823.256.750 (XO: 0.823260567)
   Min dTM/dt [1/us]:-2.27e+03 at an offset of 108.425 usec -> 2021-08-16T23:35:19.823.254.150 (XO: 0.823257967)

 - Block number: 2 starting at 2021-08-16T23:35:36.652.673.725 (XO: 17.652677079 - 17.652983080, t=37.2 degC)  Rank: 00079
   TM values: Max =   204  Min =  -254  Ave =    -4.52  Std =    20.18
   Min TM value:-254 at an offset of 123.738 usec -> 2021-08-16T23:35:36.652.797.462 (XO: 17.652800817)
   Max TM value:204 at an offset of 126.338 usec -> 2021-08-16T23:35:36.652.800.062 (XO: 17.652803417)
   Max dTM/dt [1/us]:1.58e+03 at an offset of 126.200 usec -> 2021-08-16T23:35:36.652.799.925 (XO: 17.652803279)
   Min dTM/dt [1/us]:-1.89e+03 at an offset of 123.563 usec -> 2021-08-16T23:35:36.652.797.287 (XO: 17.652800642)
