-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Dec 18 15:26:10 2021 UTC
-- File 20210816_220446_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.393663 to 29.553689
-each separately from 2s to 29s and from 393663us to 553689us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.159448 in 1 packets: 18
MIU packet time from 2021-08-16T22:04:51.393.663.000 to 2021-08-16T22:05:18.553.689.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.159448 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.6698459 to 30.6697564
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      26.9999105 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210816_220446_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.0635096 to 4.0637986
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 27.1593590  -- OK for SL --
   MIU time from 2021-08-16T22:04:51.393.663.000 to 2021-08-16T22:04:51.393.952.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 27.159448

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.2231576 to 31.2234466
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-16T22:05:18.553.400.000 to 2021-08-16T22:05:18.553.689.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.063509600 corrected by 40bit 10MHz counter of 3.669845900 obtained at 1Hz counter of 2. Result: 2.393664999
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-16T22:04:51.393.522.725
   Max 1st sample MIU time among all SL packets: 2021-08-16T22:05:18.553.548.725
   Min 1st sample XO time among all SL packets: 2.393524724
   Max 1st sample XO time among all SL packets: 29.553551752
   TM values from -248 to 247 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-16T22:04:51.393.522.725 (XO: 2.393524724 - 2.393830725, t=37.2 degC)  Rank: 00080
   TM values: Max =   212  Min =  -248  Ave =    -4.49  Std =    27.91
   Min TM value:-248 at an offset of 165.901 usec -> 2021-08-16T22:04:51.393.688.625 (XO: 2.393690624)
   Max TM value:212 at an offset of 122.938 usec -> 2021-08-16T22:04:51.393.645.662 (XO: 2.393647662)
   Max dTM/dt [1/us]:1.74e+03 at an offset of 163.488 usec -> 2021-08-16T22:04:51.393.686.213 (XO: 2.393688212)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 165.813 usec -> 2021-08-16T22:04:51.393.688.538 (XO: 2.393690537)

 - Block number: 2 starting at 2021-08-16T22:05:18.553.259.725 (XO: 29.553262751 - 29.553568752, t=37.2 degC)  Rank: 00084
   TM values: Max =   247  Min =  -246  Ave =    -4.43  Std =    21.52
   Min TM value:-246 at an offset of 122.375 usec -> 2021-08-16T22:05:18.553.382.100 (XO: 29.553385126)
   Max TM value:247 at an offset of 122.050 usec -> 2021-08-16T22:05:18.553.381.775 (XO: 29.553384801)
   Max dTM/dt [1/us]:2.36e+03 at an offset of 124.675 usec -> 2021-08-16T22:05:18.553.384.400 (XO: 29.553387426)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 122.163 usec -> 2021-08-16T22:05:18.553.381.887 (XO: 29.553384914)
