-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Dec 18 09:18:58 2021 UTC
-- File 20210816_142928_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.185668 to 15.946272
-each separately from 7s to 15s and from 185668us to 946272us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.760026 in 1 packets: 18
MIU packet time from 2021-08-16T14:29:38.185.668.000 to 2021-08-16T14:29:46.946.272.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.760026 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.8973341 to 16.8973074
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999733 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210816_142928_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.0830027 to 9.0832917
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.7599990  -- OK for SL --
   MIU time from 2021-08-16T14:29:38.185.668.000 to 2021-08-16T14:29:38.185.957.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.760026

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.8432907 to 17.8435797
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-16T14:29:46.945.983.000 to 2021-08-16T14:29:46.946.272.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.083002700 corrected by 40bit 10MHz counter of 8.897334100 obtained at 1Hz counter of 7. Result: 7.185669213
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-16T14:29:38.185.527.725
   Max 1st sample MIU time among all SL packets: 2021-08-16T14:29:46.946.131.725
   Min 1st sample XO time among all SL packets: 7.185528937
   Max 1st sample XO time among all SL packets: 15.946135147
   TM values from -261 to 203 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-16T14:29:38.185.527.725 (XO: 7.185528937 - 7.185834938, t=37.2 degC)  Rank: 00085
   TM values: Max =   203  Min =  -261  Ave =    -4.51  Std =    19.76
   Min TM value:-261 at an offset of 120.638 usec -> 2021-08-16T14:29:38.185.648.362 (XO: 7.185649575)
   Max TM value:203 at an offset of 123.238 usec -> 2021-08-16T14:29:38.185.650.962 (XO: 7.185652175)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 123.050 usec -> 2021-08-16T14:29:38.185.650.775 (XO: 7.185651988)
   Min dTM/dt [1/us]:-1.81e+03 at an offset of 120.463 usec -> 2021-08-16T14:29:38.185.648.187 (XO: 7.185649400)

 - Block number: 2 starting at 2021-08-16T14:29:46.945.842.725 (XO: 15.945846146 - 15.946152147, t=37.2 degC)  Rank: 00079
   TM values: Max =   175  Min =  -253  Ave =    -4.45  Std =    22.97
   Min TM value:-253 at an offset of 106.725 usec -> 2021-08-16T14:29:46.945.949.450 (XO: 15.945952872)
   Max TM value:175 at an offset of 106.438 usec -> 2021-08-16T14:29:46.945.949.162 (XO: 15.945952584)
   Max dTM/dt [1/us]:1.43e+03 at an offset of 122.113 usec -> 2021-08-16T14:29:46.945.964.837 (XO: 15.945968259)
   Min dTM/dt [1/us]:-1.81e+03 at an offset of 106.613 usec -> 2021-08-16T14:29:46.945.949.337 (XO: 15.945952759)
