-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Dec 18 07:22:02 2021 UTC
-- File 20210816_124738_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.528318 to 55.006310
-each separately from 5s to 55s and from 6021us to 528607us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.976567 in 1 packets: 18
      24.523763 in 1 packets: 36
      12.976506 in 1 packets: 54
MIU packet time from 2021-08-16T12:47:46.528.318.000 to 2021-08-16T12:48:36.006.310.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.976567 in 1 packets: 18
      24.523763 in 1 packets: 36
      12.976506 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.7856203 to 56.7854523
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999596 in 1 packets: 18
      24.9999159 in 1 packets: 36
      12.9999565 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210816_124738_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.3139387 to 7.3142277
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.9765270  -- OK for SL --
   MIU time from 2021-08-16T12:47:46.528.318.000 to 2021-08-16T12:47:46.528.607.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.976567

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.2907547 to 19.2910437
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.5236790  -- OK for SL --
   MIU time from 2021-08-16T12:47:58.505.174.000 to 2021-08-16T12:47:58.505.463.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.523763

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.8147227 to 43.8150117
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.9764630  -- OK for SL --
   MIU time from 2021-08-16T12:48:23.029.226.000 to 2021-08-16T12:48:23.029.515.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.976506

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 56.7914747 to 56.7917637
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-16T12:48:36.006.021.000 to 2021-08-16T12:48:36.006.310.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.313938700 corrected by 40bit 10MHz counter of 6.785620300 obtained at 1Hz counter of 5. Result: 5.528320196
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-16T12:47:46.528.177.725
   Max 1st sample MIU time among all SL packets: 2021-08-16T12:48:36.006.169.725
   Min 1st sample XO time among all SL packets: 5.528179921
   Max 1st sample XO time among all SL packets: 55.006171145
   TM values from -249 to 229 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-16T12:47:46.528.177.725 (XO: 5.528179921 - 5.528485922, t=37.2 degC)  Rank: 00077
   TM values: Max =   208  Min =  -249  Ave =    -4.55  Std =    22.83
   Min TM value:-249 at an offset of 122.750 usec -> 2021-08-16T12:47:46.528.300.475 (XO: 5.528302671)
   Max TM value:208 at an offset of 125.400 usec -> 2021-08-16T12:47:46.528.303.125 (XO: 5.528305321)
   Max dTM/dt [1/us]:1.63e+03 at an offset of 125.275 usec -> 2021-08-16T12:47:46.528.303.000 (XO: 5.528305196)
   Min dTM/dt [1/us]:-1.83e+03 at an offset of 122.625 usec -> 2021-08-16T12:47:46.528.300.350 (XO: 5.528302546)

 - Block number: 2 starting at 2021-08-16T12:47:58.505.033.725 (XO: 17.505036192 - 17.505342193, t=37.2 degC)  Rank: 00077
   TM values: Max =   229  Min =  -243  Ave =    -4.47  Std =    26.18
   Min TM value:-243 at an offset of 121.800 usec -> 2021-08-16T12:47:58.505.155.525 (XO: 17.505157992)
   Max TM value:229 at an offset of 121.538 usec -> 2021-08-16T12:47:58.505.155.262 (XO: 17.505157730)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 124.175 usec -> 2021-08-16T12:47:58.505.157.900 (XO: 17.505160367)
   Min dTM/dt [1/us]:-2.16e+03 at an offset of 121.650 usec -> 2021-08-16T12:47:58.505.155.375 (XO: 17.505157842)

 - Block number: 3 starting at 2021-08-16T12:48:23.029.085.725 (XO: 42.029086724 - 42.029392725, t=37.2 degC)  Rank: 00087
   TM values: Max =   198  Min =  -243  Ave =    -4.58  Std =    29.75
   Min TM value:-243 at an offset of 165.363 usec -> 2021-08-16T12:48:23.029.251.088 (XO: 42.029252087)
   Max TM value:198 at an offset of 149.651 usec -> 2021-08-16T12:48:23.029.235.375 (XO: 42.029236374)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 149.463 usec -> 2021-08-16T12:48:23.029.235.188 (XO: 42.029236187)
   Min dTM/dt [1/us]:-2.16e+03 at an offset of 165.251 usec -> 2021-08-16T12:48:23.029.250.975 (XO: 42.029251974)

 - Block number: 4 starting at 2021-08-16T12:48:36.005.880.725 (XO: 55.005882144 - 55.006188145, t=37.2 degC)  Rank: 00084
   TM values: Max =   187  Min =  -244  Ave =    -4.47  Std =    22.21
   Min TM value:-244 at an offset of 120.313 usec -> 2021-08-16T12:48:36.006.001.037 (XO: 55.006002457)
   Max TM value:187 at an offset of 122.963 usec -> 2021-08-16T12:48:36.006.003.687 (XO: 55.006005107)
   Max dTM/dt [1/us]:1.63e+03 at an offset of 122.738 usec -> 2021-08-16T12:48:36.006.003.462 (XO: 55.006004882)
   Min dTM/dt [1/us]:-1.78e+03 at an offset of 75.375 usec -> 2021-08-16T12:48:36.005.956.100 (XO: 55.005957520)
