-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Dec 18 04:45:40 2021 UTC
-- File 20210816_100950_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.732973 to 24.202252
-each separately from 10s to 24s and from 201963us to 733262us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.468701 in 1 packets: 18
MIU packet time from 2021-08-16T10:10:03.732.973.000 to 2021-08-16T10:10:17.202.252.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.468701 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.8770582 to 25.8770118
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999536 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210816_100950_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.6100315 to 12.6103205
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.4686550  -- OK for SL --
   MIU time from 2021-08-16T10:10:03.732.973.000 to 2021-08-16T10:10:03.733.262.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.468701

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.0789755 to 26.0792645
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-16T10:10:17.201.963.000 to 2021-08-16T10:10:17.202.252.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.610031500 corrected by 40bit 10MHz counter of 11.877058200 obtained at 1Hz counter of 10. Result: 10.732975719
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-16T10:10:03.732.832.725
   Max 1st sample MIU time among all SL packets: 2021-08-16T10:10:17.202.111.725
   Min 1st sample XO time among all SL packets: 10.732835443
   Max 1st sample XO time among all SL packets: 24.202113092
   TM values from -254 to 212 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-16T10:10:03.732.832.725 (XO: 10.732835443 - 10.733141444, t=37.2 degC)  Rank: 00083
   TM values: Max =   199  Min =  -238  Ave =    -4.48  Std =    21.82
   Min TM value:-238 at an offset of 107.875 usec -> 2021-08-16T10:10:03.732.940.600 (XO: 10.732943319)
   Max TM value:199 at an offset of 110.500 usec -> 2021-08-16T10:10:03.732.943.225 (XO: 10.732945944)
   Max dTM/dt [1/us]:1.65e+03 at an offset of 110.338 usec -> 2021-08-16T10:10:03.732.943.062 (XO: 10.732945781)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 107.688 usec -> 2021-08-16T10:10:03.732.940.412 (XO: 10.732943131)

 - Block number: 2 starting at 2021-08-16T10:10:17.201.822.725 (XO: 24.201824091 - 24.202130092, t=37.2 degC)  Rank: 00079
   TM values: Max =   212  Min =  -254  Ave =    -4.48  Std =    20.97
   Min TM value:-254 at an offset of 121.063 usec -> 2021-08-16T10:10:17.201.943.787 (XO: 24.201945154)
   Max TM value:212 at an offset of 120.763 usec -> 2021-08-16T10:10:17.201.943.487 (XO: 24.201944854)
   Max dTM/dt [1/us]:1.77e+03 at an offset of 118.638 usec -> 2021-08-16T10:10:17.201.941.362 (XO: 24.201942729)
   Min dTM/dt [1/us]:-1.94e+03 at an offset of 120.875 usec -> 2021-08-16T10:10:17.201.943.600 (XO: 24.201944966)
