-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Dec 14 04:32:23 2021 UTC
-- File 20210813_082138_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.225186 to 32.869066
-each separately from 6s to 32s and from 225186us to 911685us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.685921 in 1 packets: 18
      10.957092 in 1 packets: 36
MIU packet time from 2021-08-13T08:21:47.225.186.000 to 2021-08-13T08:22:13.869.066.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.685921 in 1 packets: 18
      10.957092 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9501693 to 33.9500824
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999499 in 1 packets: 18
      10.9999632 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210813_082138_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.1753562 to 8.1756452
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.6858710  -- OK for SL --
   MIU time from 2021-08-13T08:21:47.225.186.000 to 2021-08-13T08:21:47.225.475.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.685921

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.8615162 to 23.8618052
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.9570550  -- OK for SL --
   MIU time from 2021-08-13T08:22:02.911.396.000 to 2021-08-13T08:22:02.911.685.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.957092

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.8188602 to 34.8191492
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-13T08:22:13.868.777.000 to 2021-08-13T08:22:13.869.066.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.175356200 corrected by 40bit 10MHz counter of 7.950169300 obtained at 1Hz counter of 6. Result: 6.225187666
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-13T08:21:47.225.045.725
   Max 1st sample MIU time among all SL packets: 2021-08-13T08:22:13.868.925.725
   Min 1st sample XO time among all SL packets: 6.225047390
   Max 1st sample XO time among all SL packets: 32.868929392
   TM values from -280 to 248 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-13T08:21:47.225.045.725 (XO: 6.225047390 - 6.225353391, t=37.2 degC)  Rank: 00080
   TM values: Max =   230  Min =  -268  Ave =    -4.42  Std =    25.32
   Min TM value:-268 at an offset of 97.850 usec -> 2021-08-13T08:21:47.225.143.575 (XO: 6.225145240)
   Max TM value:230 at an offset of 124.825 usec -> 2021-08-13T08:21:47.225.170.550 (XO: 6.225172216)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 122.750 usec -> 2021-08-13T08:21:47.225.168.475 (XO: 6.225170141)
   Min dTM/dt [1/us]:-2.22e+03 at an offset of 124.950 usec -> 2021-08-13T08:21:47.225.170.675 (XO: 6.225172341)

 - Block number: 2 starting at 2021-08-13T08:22:02.911.255.725 (XO: 21.911259823 - 21.911565824, t=37.2 degC)  Rank: 00082
   TM values: Max =   221  Min =  -280  Ave =    -4.54  Std =    34.95
   Min TM value:-280 at an offset of 222.263 usec -> 2021-08-13T08:22:02.911.477.988 (XO: 21.911482087)
   Max TM value:221 at an offset of 122.463 usec -> 2021-08-13T08:22:02.911.378.187 (XO: 21.911382286)
   Max dTM/dt [1/us]:2.19e+03 at an offset of 125.100 usec -> 2021-08-13T08:22:02.911.380.825 (XO: 21.911384924)
   Min dTM/dt [1/us]:-2.32e+03 at an offset of 222.176 usec -> 2021-08-13T08:22:02.911.477.900 (XO: 21.911481999)

 - Block number: 3 starting at 2021-08-13T08:22:13.868.636.725 (XO: 32.868640392 - 32.868946393, t=37.2 degC)  Rank: 00078
   TM values: Max =   248  Min =  -271  Ave =    -4.47  Std =    27.98
   Min TM value:-271 at an offset of 154.913 usec -> 2021-08-13T08:22:13.868.791.638 (XO: 32.868795305)
   Max TM value:248 at an offset of 113.950 usec -> 2021-08-13T08:22:13.868.750.675 (XO: 32.868754342)
   Max dTM/dt [1/us]:2.05e+03 at an offset of 116.600 usec -> 2021-08-13T08:22:13.868.753.325 (XO: 32.868756992)
   Min dTM/dt [1/us]:-2.57e+03 at an offset of 114.050 usec -> 2021-08-13T08:22:13.868.750.775 (XO: 32.868754442)
