-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Dec 14 04:03:40 2021 UTC
-- File 20210813_075158_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.165970 to 46.395849
-each separately from 1s to 46s and from 165970us to 980510us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.596808 in 1 packets: 18
      15.216865 in 1 packets: 36
      12.415050 in 1 packets: 54
MIU packet time from 2021-08-13T07:52:02.165.970.000 to 2021-08-13T07:52:47.395.849.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.596808 in 1 packets: 18
      15.216865 in 1 packets: 36
      12.415050 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.8976820 to 47.8975315
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999432 in 1 packets: 18
      14.9999499 in 1 packets: 36
      12.9999564 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      03 -> 51.1 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 53 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 53 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210813_075158_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.0636526 to 3.0639416
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.5967510  -- OK for SL --
   MIU time from 2021-08-13T07:52:02.165.970.000 to 2021-08-13T07:52:02.166.259.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.596808

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.6606926 to 20.6609816
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.2168150  -- OK for SL --
   MIU time from 2021-08-13T07:52:19.763.067.000 to 2021-08-13T07:52:19.763.356.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.216865

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.8777966 to 35.8780856
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.4150070  -- OK for SL --
   MIU time from 2021-08-13T07:52:34.980.221.000 to 2021-08-13T07:52:34.980.510.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.415050

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 48.2930926 to 48.2933816
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-13T07:52:47.395.560.000 to 2021-08-13T07:52:47.395.849.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.063652600 corrected by 40bit 10MHz counter of 2.897682000 obtained at 1Hz counter of 1. Result: 1.165971164
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-13T07:52:02.165.829.725
   Max 1st sample MIU time among all SL packets: 2021-08-13T07:52:47.395.708.725
   Min 1st sample XO time among all SL packets: 1.165830889
   Max 1st sample XO time among all SL packets: 46.395711131
   TM values from -276 to 244 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-13T07:52:02.165.829.725 (XO: 1.165830889 - 1.166136890, t=37.2 degC)  Rank: 00077
   TM values: Max =   244  Min =  -261  Ave =    -4.43  Std =    19.17
   Min TM value:-261 at an offset of 93.025 usec -> 2021-08-13T07:52:02.165.922.750 (XO: 1.165923914)
   Max TM value:244 at an offset of 92.788 usec -> 2021-08-13T07:52:02.165.922.512 (XO: 1.165923677)
   Max dTM/dt [1/us]:2.23e+03 at an offset of 95.425 usec -> 2021-08-13T07:52:02.165.925.150 (XO: 1.165926314)
   Min dTM/dt [1/us]:-2.61e+03 at an offset of 92.900 usec -> 2021-08-13T07:52:02.165.922.625 (XO: 1.165923789)

 - Block number: 2 starting at 2021-08-13T07:52:19.762.926.725 (XO: 18.762929643 - 18.763235644, t=37.2 degC)  Rank: 00081
   TM values: Max =   217  Min =  -253  Ave =    -4.50  Std =    27.10
   Min TM value:-253 at an offset of 101.600 usec -> 2021-08-13T07:52:19.763.028.325 (XO: 18.763031243)
   Max TM value:217 at an offset of 101.300 usec -> 2021-08-13T07:52:19.763.028.025 (XO: 18.763030943)
   Max dTM/dt [1/us]:1.64e+03 at an offset of 161.751 usec -> 2021-08-13T07:52:19.763.088.475 (XO: 18.763091393)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 101.413 usec -> 2021-08-13T07:52:19.763.028.137 (XO: 18.763031056)

 - Block number: 3 starting at 2021-08-13T07:52:34.980.080.725 (XO: 33.980084557 - 33.980390558, t=37.2 degC)  Rank: 00080
   TM values: Max =   228  Min =  -276  Ave =    -4.53  Std =    34.00
   Min TM value:-276 at an offset of 200.676 usec -> 2021-08-13T07:52:34.980.281.400 (XO: 33.980285233)
   Max TM value:228 at an offset of 131.113 usec -> 2021-08-13T07:52:34.980.211.837 (XO: 33.980215670)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 133.750 usec -> 2021-08-13T07:52:34.980.214.475 (XO: 33.980218308)
   Min dTM/dt [1/us]:-2.36e+03 at an offset of 214.063 usec -> 2021-08-13T07:52:34.980.294.788 (XO: 33.980298621)

 - Block number: 4 starting at 2021-08-13T07:52:47.395.419.725 (XO: 46.395422130 - 46.395728131, t=37.2 degC)  Rank: 00077
   TM values: Max =   220  Min =  -267  Ave =    -4.37  Std =    25.19
   Min TM value:-267 at an offset of 128.425 usec -> 2021-08-13T07:52:47.395.548.150 (XO: 46.395550555)
   Max TM value:220 at an offset of 128.125 usec -> 2021-08-13T07:52:47.395.547.850 (XO: 46.395550255)
   Max dTM/dt [1/us]:2.17e+03 at an offset of 130.788 usec -> 2021-08-13T07:52:47.395.550.512 (XO: 46.395552918)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 128.238 usec -> 2021-08-13T07:52:47.395.547.962 (XO: 46.395550368)
