-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Dec 14 03:10:34 2021 UTC
-- File 20210813_053418_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.070037 to 43.821071
-each separately from 7s to 43s and from 70037us to 821071us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.160281 in 1 packets: 18
      14.589886 in 1 packets: 36
MIU packet time from 2021-08-13T05:34:28.070.037.000 to 2021-08-13T05:35:04.821.071.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.160281 in 1 packets: 18
      14.589886 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.8779570 to 44.8778371
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999267 in 1 packets: 18
      13.9999534 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 1 packets: 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 1 packets: 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210813_053418_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      70 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.9479955 to 8.9482845
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.1602070  -- OK for SL --
   MIU time from 2021-08-13T05:34:28.070.037.000 to 2021-08-13T05:34:28.070.326.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.160281

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.1084915 to 31.1087805
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.5898390  -- OK for SL --
   MIU time from 2021-08-13T05:34:50.230.607.000 to 2021-08-13T05:34:50.230.896.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.589886

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.6986195 to 45.6989085
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-13T05:35:04.820.782.000 to 2021-08-13T05:35:04.821.071.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 8.947995500 corrected by 40bit 10MHz counter of 8.877957000 obtained at 1Hz counter of 7. Result: 7.070038731
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-13T05:34:28.069.896.725
   Max 1st sample MIU time among all SL packets: 2021-08-13T05:35:04.820.930.725
   Min 1st sample XO time among all SL packets: 7.069898456
   Max 1st sample XO time among all SL packets: 43.820933834
   TM values from -268 to 241 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-13T05:34:28.069.896.725 (XO: 7.069898456 - 7.070204457, t=37.2 degC)  Rank: 00070
   TM values: Max =   206  Min =  -268  Ave =    -4.54  Std =    23.39
   Min TM value:-268 at an offset of 118.875 usec -> 2021-08-13T05:34:28.070.015.600 (XO: 7.070017331)
   Max TM value:206 at an offset of 118.575 usec -> 2021-08-13T05:34:28.070.015.300 (XO: 7.070017031)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 121.263 usec -> 2021-08-13T05:34:28.070.017.987 (XO: 7.070019719)
   Min dTM/dt [1/us]:-2.14e+03 at an offset of 118.663 usec -> 2021-08-13T05:34:28.070.015.387 (XO: 7.070017119)

 - Block number: 2 starting at 2021-08-13T05:34:50.230.466.725 (XO: 29.230468309 - 29.230774310, t=37.2 degC)  Rank: 00072
   TM values: Max =   218  Min =  -268  Ave =    -4.35  Std =    25.20
   Min TM value:-268 at an offset of 115.275 usec -> 2021-08-13T05:34:50.230.582.000 (XO: 29.230583584)
   Max TM value:218 at an offset of 115.000 usec -> 2021-08-13T05:34:50.230.581.725 (XO: 29.230583309)
   Max dTM/dt [1/us]:2.43e+03 at an offset of 117.638 usec -> 2021-08-13T05:34:50.230.584.362 (XO: 29.230585946)
   Min dTM/dt [1/us]:-2.32e+03 at an offset of 115.050 usec -> 2021-08-13T05:34:50.230.581.775 (XO: 29.230583359)

 - Block number: 3 starting at 2021-08-13T05:35:04.820.641.725 (XO: 43.820644833 - 43.820950834, t=37.2 degC)  Rank: 00086
   TM values: Max =   241  Min =  -235  Ave =    -4.42  Std =    18.43
   Min TM value:-235 at an offset of 132.875 usec -> 2021-08-13T05:35:04.820.774.600 (XO: 43.820777709)
   Max TM value:241 at an offset of 130.625 usec -> 2021-08-13T05:35:04.820.772.350 (XO: 43.820775459)
   Max dTM/dt [1/us]:2.09e+03 at an offset of 135.225 usec -> 2021-08-13T05:35:04.820.776.950 (XO: 43.820780059)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 132.663 usec -> 2021-08-13T05:35:04.820.774.387 (XO: 43.820777496)
