-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Dec 14 02:41:48 2021 UTC
-- File 20210813_051819_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.076219 to 39.056617
-each separately from 11s to 39s and from 56328us to 646679us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.569882 in 1 packets: 18
      14.409649 in 1 packets: 36
MIU packet time from 2021-08-13T05:18:33.076.219.000 to 2021-08-13T05:19:01.056.617.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.569882 in 1 packets: 18
      14.409649 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.8724097 to 40.8723170
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999570 in 1 packets: 18
      14.9999503 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      06 -> 47.2 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210813_051819_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.9486299 to 12.9489189
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.5698390  -- OK for SL --
   MIU time from 2021-08-13T05:18:33.076.219.000 to 2021-08-13T05:18:33.076.508.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.569882

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.5187579 to 26.5190469
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.4095990  -- OK for SL --
   MIU time from 2021-08-13T05:18:46.646.390.000 to 2021-08-13T05:18:46.646.679.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.409649

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.9286459 to 40.9289349
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-13T05:19:01.056.328.000 to 2021-08-13T05:19:01.056.617.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 12.948629900 corrected by 40bit 10MHz counter of 12.872409700 obtained at 1Hz counter of 11. Result: 11.076220444
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-08-13T05:18:33.076.078.725
   Max 1st sample MIU time among all SL packets: 2021-08-13T05:19:01.056.476.725
   Min 1st sample XO time among all SL packets: 11.076080168
   Max 1st sample XO time among all SL packets: 39.056477811
   TM values from -277 to 217 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-13T05:18:33.076.078.725 (XO: 11.076080168 - 11.076386169, t=37.2 degC)  Rank: 00078
   TM values: Max =   213  Min =  -277  Ave =    -4.48  Std =    29.96
   Min TM value:-277 at an offset of 215.826 usec -> 2021-08-13T05:18:33.076.294.550 (XO: 11.076295994)
   Max TM value:213 at an offset of 129.538 usec -> 2021-08-13T05:18:33.076.208.262 (XO: 11.076209706)
   Max dTM/dt [1/us]:2.03e+03 at an offset of 132.200 usec -> 2021-08-13T05:18:33.076.210.925 (XO: 11.076212369)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 215.726 usec -> 2021-08-13T05:18:33.076.294.450 (XO: 11.076295894)

 - Block number: 2 starting at 2021-08-13T05:18:46.646.249.725 (XO: 24.646253058 - 24.646559059, t=37.2 degC)  Rank: 00083
   TM values: Max =   189  Min =  -226  Ave =    -4.42  Std =    24.69
   Min TM value:-226 at an offset of 111.950 usec -> 2021-08-13T05:18:46.646.361.675 (XO: 24.646365008)
   Max TM value:189 at an offset of 128.013 usec -> 2021-08-13T05:18:46.646.377.737 (XO: 24.646381071)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 127.875 usec -> 2021-08-13T05:18:46.646.377.600 (XO: 24.646380933)
   Min dTM/dt [1/us]:-1.70e+03 at an offset of 80.050 usec -> 2021-08-13T05:18:46.646.329.775 (XO: 24.646333108)

 - Block number: 3 starting at 2021-08-13T05:19:01.056.187.725 (XO: 39.056188810 - 39.056494811, t=37.2 degC)  Rank: 00082
   TM values: Max =   217  Min =  -261  Ave =    -4.49  Std =    21.74
   Min TM value:-261 at an offset of 127.663 usec -> 2021-08-13T05:19:01.056.315.387 (XO: 39.056316473)
   Max TM value:217 at an offset of 130.200 usec -> 2021-08-13T05:19:01.056.317.925 (XO: 39.056319011)
   Max dTM/dt [1/us]:2.07e+03 at an offset of 130.025 usec -> 2021-08-13T05:19:01.056.317.750 (XO: 39.056318836)
   Min dTM/dt [1/us]:-2.13e+03 at an offset of 127.450 usec -> 2021-08-13T05:19:01.056.315.175 (XO: 39.056316261)
