-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Dec 14 00:25:49 2021 UTC
-- File 20210813_032759_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.303605 to 19.953948
-each separately from 3s to 19s and from 303605us to 953948us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.649765 in 1 packets: 18
MIU packet time from 2021-08-13T03:28:05.303.605.000 to 2021-08-13T03:28:21.953.948.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.649765 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.8422845 to 20.8422312
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999467 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210813_032759_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.1458908 to 5.1461798
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.6497110  -- OK for SL --
   MIU time from 2021-08-13T03:28:05.303.605.000 to 2021-08-13T03:28:05.303.894.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.649765

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.7958908 to 21.7961798
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-13T03:28:21.953.659.000 to 2021-08-13T03:28:21.953.948.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.145890800 corrected by 40bit 10MHz counter of 4.842284500 obtained at 1Hz counter of 3. Result: 3.303607302
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-13T03:28:05.303.464.725
   Max 1st sample MIU time among all SL packets: 2021-08-13T03:28:21.953.807.725
   Min 1st sample XO time among all SL packets: 3.303467026
   Max 1st sample XO time among all SL packets: 19.953811473
   TM values from -250 to 237 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-13T03:28:05.303.464.725 (XO: 3.303467026 - 3.303773027, t=37.2 degC)  Rank: 00080
   TM values: Max =   237  Min =  -250  Ave =    -4.41  Std =    17.83
   Min TM value:-250 at an offset of 106.713 usec -> 2021-08-13T03:28:05.303.571.437 (XO: 3.303573739)
   Max TM value:237 at an offset of 104.438 usec -> 2021-08-13T03:28:05.303.569.162 (XO: 3.303571464)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 104.325 usec -> 2021-08-13T03:28:05.303.569.050 (XO: 3.303571352)
   Min dTM/dt [1/us]:-1.95e+03 at an offset of 106.575 usec -> 2021-08-13T03:28:05.303.571.300 (XO: 3.303573602)

 - Block number: 2 starting at 2021-08-13T03:28:21.953.518.725 (XO: 19.953522472 - 19.953828473, t=37.2 degC)  Rank: 00080
   TM values: Max =   230  Min =  -247  Ave =    -4.47  Std =    21.05
   Min TM value:-247 at an offset of 127.450 usec -> 2021-08-13T03:28:21.953.646.175 (XO: 19.953649922)
   Max TM value:230 at an offset of 130.050 usec -> 2021-08-13T03:28:21.953.648.775 (XO: 19.953652522)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 129.888 usec -> 2021-08-13T03:28:21.953.648.612 (XO: 19.953652360)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 127.213 usec -> 2021-08-13T03:28:21.953.645.937 (XO: 19.953649685)
