-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Dec 12 08:45:16 2021 UTC
-- File 20210811_142613_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.992623 to 69.012174
-each separately from 12s to 69s and from 11885us to 992912us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.019540 in 1 packets: 18
      2.919654 in 1 packets: 36
      22.439547 in 1 packets: 54
      15.639365 in 1 packets: 72
MIU packet time from 2021-08-11T14:26:28.992.623.000 to 2021-08-11T14:27:25.012.174.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.019540 in 1 packets: 18
      2.919654 in 1 packets: 36
      22.439547 in 1 packets: 54
      15.639365 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.8648058 to 70.8646153
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999465 in 2 packets: 18, 72
      1.9999934 in 1 packets: 36
      22.9999231 in 1 packets: 54
Temperature FPGA: 
      05 -> 48.5 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 54 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 54 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210811_142613_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.8574291 to 14.8577181
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0194870  -- OK for SL --
   MIU time from 2021-08-11T14:26:28.992.623.000 to 2021-08-11T14:26:28.992.912.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.019540

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.8772051 to 29.8774941
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 2.9196470  -- OK for SL --
   MIU time from 2021-08-11T14:26:44.012.452.000 to 2021-08-11T14:26:44.012.741.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 2.919654

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.7971411 to 32.7974301
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.4394710  -- OK for SL --
   MIU time from 2021-08-11T14:26:46.932.395.000 to 2021-08-11T14:26:46.932.684.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.439547

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 55.2369011 to 55.2371901
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.6393110  -- OK for SL --
   MIU time from 2021-08-11T14:27:09.372.231.000 to 2021-08-11T14:27:09.372.520.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.639365

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 70.8765011 to 70.8767901
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-11T14:27:25.011.885.000 to 2021-08-11T14:27:25.012.174.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.857429100 corrected by 40bit 10MHz counter of 13.864805800 obtained at 1Hz counter of 12. Result: 12.992626576
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-11T14:26:28.992.482.725
   Max 1st sample MIU time among all SL packets: 2021-08-11T14:27:25.012.033.725
   Min 1st sample XO time among all SL packets: 12.992486300
   Max 1st sample XO time among all SL packets: 69.012034565
   TM values from -254 to 228 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-11T14:26:28.992.482.725 (XO: 12.992486300 - 12.992792301, t=37.2 degC)  Rank: 00080
   TM values: Max =   196  Min =  -244  Ave =    -4.42  Std =    18.97
   Min TM value:-244 at an offset of 121.450 usec -> 2021-08-11T14:26:28.992.604.175 (XO: 12.992607751)
   Max TM value:196 at an offset of 121.150 usec -> 2021-08-11T14:26:28.992.603.875 (XO: 12.992607451)
   Max dTM/dt [1/us]:1.78e+03 at an offset of 123.913 usec -> 2021-08-11T14:26:28.992.606.637 (XO: 12.992610213)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 121.263 usec -> 2021-08-11T14:26:28.992.603.987 (XO: 12.992607563)

 - Block number: 2 starting at 2021-08-11T14:26:44.012.311.725 (XO: 28.012312566 - 28.012618567, t=37.2 degC)  Rank: 00085
   TM values: Max =   215  Min =  -251  Ave =    -4.43  Std =    19.92
   Min TM value:-251 at an offset of 105.888 usec -> 2021-08-11T14:26:44.012.417.612 (XO: 28.012418453)
   Max TM value:215 at an offset of 105.600 usec -> 2021-08-11T14:26:44.012.417.325 (XO: 28.012418166)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 108.325 usec -> 2021-08-11T14:26:44.012.420.050 (XO: 28.012420891)
   Min dTM/dt [1/us]:-2.19e+03 at an offset of 105.700 usec -> 2021-08-11T14:26:44.012.417.425 (XO: 28.012418266)

 - Block number: 3 starting at 2021-08-11T14:26:46.932.254.725 (XO: 30.932258295 - 30.932564296, t=37.2 degC)  Rank: 00077
   TM values: Max =   206  Min =  -247  Ave =    -4.35  Std =    22.51
   Min TM value:-247 at an offset of 120.600 usec -> 2021-08-11T14:26:46.932.375.325 (XO: 30.932378895)
   Max TM value:206 at an offset of 123.225 usec -> 2021-08-11T14:26:46.932.377.950 (XO: 30.932381520)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 123.025 usec -> 2021-08-11T14:26:46.932.377.750 (XO: 30.932381320)
   Min dTM/dt [1/us]:-2.32e+03 at an offset of 120.388 usec -> 2021-08-11T14:26:46.932.375.112 (XO: 30.932378683)

 - Block number: 4 starting at 2021-08-11T14:27:09.372.090.725 (XO: 53.372093253 - 53.372399254, t=37.2 degC)  Rank: 00080
   TM values: Max =   228  Min =  -254  Ave =    -4.44  Std =    32.33
   Min TM value:-254 at an offset of 167.401 usec -> 2021-08-11T14:27:09.372.258.125 (XO: 53.372260653)
   Max TM value:228 at an offset of 110.688 usec -> 2021-08-11T14:27:09.372.201.412 (XO: 53.372203941)
   Max dTM/dt [1/us]:1.97e+03 at an offset of 113.363 usec -> 2021-08-11T14:27:09.372.204.087 (XO: 53.372206616)
   Min dTM/dt [1/us]:-2.46e+03 at an offset of 110.800 usec -> 2021-08-11T14:27:09.372.201.525 (XO: 53.372204053)

 - Block number: 5 starting at 2021-08-11T14:27:25.011.744.725 (XO: 69.011745564 - 69.012051565, t=37.2 degC)  Rank: 00081
   TM values: Max =   211  Min =  -252  Ave =    -4.44  Std =    20.61
   Min TM value:-252 at an offset of 118.225 usec -> 2021-08-11T14:27:25.011.862.950 (XO: 69.011863789)
   Max TM value:211 at an offset of 117.900 usec -> 2021-08-11T14:27:25.011.862.625 (XO: 69.011863464)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 120.588 usec -> 2021-08-11T14:27:25.011.865.312 (XO: 69.011866152)
   Min dTM/dt [1/us]:-1.85e+03 at an offset of 118.000 usec -> 2021-08-11T14:27:25.011.862.725 (XO: 69.011863564)
