-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Dec 12 00:51:56 2021 UTC
-- File 20210810_160012_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.535842 to 22.615992
-each separately from 6s to 22s and from 535842us to 615992us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.079572 in 1 packets: 18
MIU packet time from 2021-08-10T16:00:21.535.842.000 to 2021-08-10T16:00:37.615.992.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.079572 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8365679 to 23.8365143
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999464 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      3 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      3 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210810_160012_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.3724105 to 8.3726995
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.0795190  -- OK for SL --
   MIU time from 2021-08-10T16:00:21.535.842.000 to 2021-08-10T16:00:21.536.131.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.079572

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.4522185 to 24.4525075
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-10T16:00:37.615.703.000 to 2021-08-10T16:00:37.615.992.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.372410500 corrected by 40bit 10MHz counter of 7.836567900 obtained at 1Hz counter of 6. Result: 6.535844422
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-10T16:00:21.535.701.725
   Max 1st sample MIU time among all SL packets: 2021-08-10T16:00:37.615.851.725
   Min 1st sample XO time among all SL packets: 6.535704146
   Max 1st sample XO time among all SL packets: 22.615854957
   TM values from -260 to 236 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-10T16:00:21.535.701.725 (XO: 6.535704146 - 6.536010147, t=37.2 degC)  Rank: 00084
   TM values: Max =   228  Min =  -235  Ave =    -4.38  Std =    19.30
   Min TM value:-235 at an offset of 109.438 usec -> 2021-08-10T16:00:21.535.811.162 (XO: 6.535813584)
   Max TM value:228 at an offset of 112.088 usec -> 2021-08-10T16:00:21.535.813.812 (XO: 6.535816234)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 111.913 usec -> 2021-08-10T16:00:21.535.813.637 (XO: 6.535816059)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 109.238 usec -> 2021-08-10T16:00:21.535.810.962 (XO: 6.535813384)

 - Block number: 2 starting at 2021-08-10T16:00:37.615.562.725 (XO: 22.615565956 - 22.615871957, t=37.2 degC)  Rank: 00089
   TM values: Max =   236  Min =  -260  Ave =    -4.40  Std =    22.87
   Min TM value:-260 at an offset of 121.263 usec -> 2021-08-10T16:00:37.615.683.987 (XO: 22.615687219)
   Max TM value:236 at an offset of 120.963 usec -> 2021-08-10T16:00:37.615.683.687 (XO: 22.615686919)
   Max dTM/dt [1/us]:2.26e+03 at an offset of 123.650 usec -> 2021-08-10T16:00:37.615.686.375 (XO: 22.615689607)
   Min dTM/dt [1/us]:-2.46e+03 at an offset of 121.063 usec -> 2021-08-10T16:00:37.615.683.787 (XO: 22.615687019)
