-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Dec 11 12:34:56 2021 UTC
-- File 20210810_095628_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.087077 to 27.559190
-each separately from 3s to 27s and from 87077us to 559190us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.471535 in 1 packets: 18
MIU packet time from 2021-08-10T09:56:34.087.077.000 to 2021-08-10T09:56:58.559.190.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.471535 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.4404121 to 28.4403324
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.9999203 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210810_095628_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.5274902 to 4.5277792
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.4714550  -- OK for SL --
   MIU time from 2021-08-10T09:56:34.087.077.000 to 2021-08-10T09:56:34.087.366.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.471535

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.9992342 to 28.9995232
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-10T09:56:58.558.901.000 to 2021-08-10T09:56:58.559.190.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 4.527490200 corrected by 40bit 10MHz counter of 4.440412100 obtained at 1Hz counter of 3. Result: 3.087078387
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-10T09:56:34.086.936.725
   Max 1st sample MIU time among all SL packets: 2021-08-10T09:56:58.559.049.725
   Min 1st sample XO time among all SL packets: 3.086938112
   Max 1st sample XO time among all SL packets: 27.559052370
   TM values from -271 to 228 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-10T09:56:34.086.936.725 (XO: 3.086938112 - 3.087244113, t=35.6 degC)  Rank: 00080
   TM values: Max =   228  Min =  -268  Ave =    -4.52  Std =    24.00
   Min TM value:-268 at an offset of 123.750 usec -> 2021-08-10T09:56:34.087.060.475 (XO: 3.087061862)
   Max TM value:228 at an offset of 121.438 usec -> 2021-08-10T09:56:34.087.058.162 (XO: 3.087059550)
   Max dTM/dt [1/us]:2.07e+03 at an offset of 121.313 usec -> 2021-08-10T09:56:34.087.058.037 (XO: 3.087059425)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 97.563 usec -> 2021-08-10T09:56:34.087.034.287 (XO: 3.087035675)

 - Block number: 2 starting at 2021-08-10T09:56:58.558.760.725 (XO: 27.558763369 - 27.559069370, t=35.6 degC)  Rank: 00082
   TM values: Max =   213  Min =  -271  Ave =    -4.52  Std =    21.87
   Min TM value:-271 at an offset of 111.425 usec -> 2021-08-10T09:56:58.558.872.150 (XO: 27.558874794)
   Max TM value:213 at an offset of 111.125 usec -> 2021-08-10T09:56:58.558.871.850 (XO: 27.558874494)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 108.938 usec -> 2021-08-10T09:56:58.558.869.662 (XO: 27.558872307)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 111.225 usec -> 2021-08-10T09:56:58.558.871.950 (XO: 27.558874594)
