-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Dec 11 11:31:36 2021 UTC
-- File 20210810_091010_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.884728 to 29.825180
-each separately from 4s to 29s and from 824891us to 885017us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.939874 in 1 packets: 18
MIU packet time from 2021-08-10T09:10:17.884.728.000 to 2021-08-10T09:10:42.825.180.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.939874 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.8762050 to 30.8761221
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.9999171 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210810_091010_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.7609337 to 6.7612227
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.9397910  -- OK for SL --
   MIU time from 2021-08-10T09:10:17.884.728.000 to 2021-08-10T09:10:17.885.017.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.939874

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.7010137 to 31.7013027
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-10T09:10:42.824.891.000 to 2021-08-10T09:10:42.825.180.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.760933700 corrected by 40bit 10MHz counter of 5.876205000 obtained at 1Hz counter of 4. Result: 4.884731620
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-10T09:10:17.884.587.725
   Max 1st sample MIU time among all SL packets: 2021-08-10T09:10:42.825.039.725
   Min 1st sample XO time among all SL packets: 4.884591344
   Max 1st sample XO time among all SL packets: 29.825043130
   TM values from -278 to 226 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-10T09:10:17.884.587.725 (XO: 4.884591344 - 4.884897345, t=37.2 degC)  Rank: 00086
   TM values: Max =   214  Min =  -244  Ave =    -4.48  Std =    17.60
   Min TM value:-244 at an offset of 117.150 usec -> 2021-08-10T09:10:17.884.704.875 (XO: 4.884708495)
   Max TM value:214 at an offset of 119.850 usec -> 2021-08-10T09:10:17.884.707.575 (XO: 4.884711195)
   Max dTM/dt [1/us]:1.75e+03 at an offset of 119.700 usec -> 2021-08-10T09:10:17.884.707.425 (XO: 4.884711045)
   Min dTM/dt [1/us]:-1.81e+03 at an offset of 116.938 usec -> 2021-08-10T09:10:17.884.704.662 (XO: 4.884708282)

 - Block number: 2 starting at 2021-08-10T09:10:42.824.750.725 (XO: 29.824754129 - 29.825060130, t=37.2 degC)  Rank: 00079
   TM values: Max =   226  Min =  -278  Ave =    -4.40  Std =    28.56
   Min TM value:-278 at an offset of 161.613 usec -> 2021-08-10T09:10:42.824.912.338 (XO: 29.824915742)
   Max TM value:226 at an offset of 119.900 usec -> 2021-08-10T09:10:42.824.870.625 (XO: 29.824874030)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 122.613 usec -> 2021-08-10T09:10:42.824.873.337 (XO: 29.824876742)
   Min dTM/dt [1/us]:-2.14e+03 at an offset of 120.000 usec -> 2021-08-10T09:10:42.824.870.725 (XO: 29.824874130)
