-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Dec 11 08:45:15 2021 UTC
-- File 20210810_063350_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.121393 to 20.302144
-each separately from 11s to 20s and from 121393us to 302144us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.180173 in 1 packets: 18
MIU packet time from 2021-08-10T06:34:04.121.393.000 to 2021-08-10T06:34:13.302.144.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.180173 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.7890644 to 21.7890342
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999698 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210810_063350_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.9104577 to 12.9107467
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.1801430  -- OK for SL --
   MIU time from 2021-08-10T06:34:04.121.393.000 to 2021-08-10T06:34:04.121.682.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.180173

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.0908897 to 22.0911787
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-10T06:34:13.301.855.000 to 2021-08-10T06:34:13.302.144.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 12.910457700 corrected by 40bit 10MHz counter of 12.789064400 obtained at 1Hz counter of 11. Result: 11.121393713
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-08-10T06:34:04.121.252.725
   Max 1st sample MIU time among all SL packets: 2021-08-10T06:34:13.302.003.725
   Min 1st sample XO time among all SL packets: 11.121253437
   Max 1st sample XO time among all SL packets: 20.302005252
   TM values from -280 to 239 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-10T06:34:04.121.252.725 (XO: 11.121253437 - 11.121559438, t=37.2 degC)  Rank: 00086
   TM values: Max =   198  Min =  -280  Ave =    -4.55  Std =    25.23
   Min TM value:-280 at an offset of 125.125 usec -> 2021-08-10T06:34:04.121.377.850 (XO: 11.121378563)
   Max TM value:198 at an offset of 127.650 usec -> 2021-08-10T06:34:04.121.380.375 (XO: 11.121381088)
   Max dTM/dt [1/us]:2.00e+03 at an offset of 127.525 usec -> 2021-08-10T06:34:04.121.380.250 (XO: 11.121380963)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 124.938 usec -> 2021-08-10T06:34:04.121.377.662 (XO: 11.121378375)

 - Block number: 2 starting at 2021-08-10T06:34:13.301.714.725 (XO: 20.301716251 - 20.302022252, t=37.2 degC)  Rank: 00074
   TM values: Max =   239  Min =  -268  Ave =    -4.54  Std =    22.70
   Min TM value:-268 at an offset of 115.675 usec -> 2021-08-10T06:34:13.301.830.400 (XO: 20.301831926)
   Max TM value:239 at an offset of 115.350 usec -> 2021-08-10T06:34:13.301.830.075 (XO: 20.301831601)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 118.000 usec -> 2021-08-10T06:34:13.301.832.725 (XO: 20.301834251)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 115.425 usec -> 2021-08-10T06:34:13.301.830.150 (XO: 20.301831676)
