-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Dec 10 04:10:05 2021 UTC
-- File 20210808_221609_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.545642 to 43.525194
-each separately from 5s to 43s and from 245346us to 545931us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.699415 in 1 packets: 18
      26.279270 in 1 packets: 36
MIU packet time from 2021-08-08T22:16:17.545.642.000 to 2021-08-08T22:16:55.525.194.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.699415 in 1 packets: 18
      26.279270 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.2637899 to 45.2636629
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999599 in 1 packets: 18
      25.9999131 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210808_221609_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.8094324 to 7.8097214
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.6993750  -- OK for SL --
   MIU time from 2021-08-08T22:16:17.545.642.000 to 2021-08-08T22:16:17.545.931.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.699415

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.5090964 to 19.5093854
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.2791830  -- OK for SL --
   MIU time from 2021-08-08T22:16:29.245.346.000 to 2021-08-08T22:16:29.245.635.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.279270

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.7885684 to 45.7888574
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-08T22:16:55.524.905.000 to 2021-08-08T22:16:55.525.194.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.809432400 corrected by 40bit 10MHz counter of 7.263789900 obtained at 1Hz counter of 5. Result: 5.545644355
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-08T22:16:17.545.501.725
   Max 1st sample MIU time among all SL packets: 2021-08-08T22:16:55.525.053.725
   Min 1st sample XO time among all SL packets: 5.545504080
   Max 1st sample XO time among all SL packets: 43.525055958
   TM values from -244 to 254 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-08T22:16:17.545.501.725 (XO: 5.545504080 - 5.545810081, t=37.2 degC)  Rank: 00080
   TM values: Max =   223  Min =  -234  Ave =    -4.46  Std =    30.25
   Min TM value:-234 at an offset of 138.288 usec -> 2021-08-08T22:16:17.545.640.012 (XO: 5.545642368)
   Max TM value:223 at an offset of 155.651 usec -> 2021-08-08T22:16:17.545.657.375 (XO: 5.545659730)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 155.476 usec -> 2021-08-08T22:16:17.545.657.200 (XO: 5.545659555)
   Min dTM/dt [1/us]:-2.15e+03 at an offset of 138.188 usec -> 2021-08-08T22:16:17.545.639.912 (XO: 5.545642268)

 - Block number: 2 starting at 2021-08-08T22:16:29.245.205.725 (XO: 17.245207159 - 17.245513160, t=37.2 degC)  Rank: 00080
   TM values: Max =   254  Min =  -244  Ave =    -4.47  Std =    22.25
   Min TM value:-244 at an offset of 119.363 usec -> 2021-08-08T22:16:29.245.325.087 (XO: 17.245326522)
   Max TM value:254 at an offset of 122.063 usec -> 2021-08-08T22:16:29.245.327.787 (XO: 17.245329222)
   Max dTM/dt [1/us]:2.13e+03 at an offset of 121.888 usec -> 2021-08-08T22:16:29.245.327.612 (XO: 17.245329047)
   Min dTM/dt [1/us]:-1.77e+03 at an offset of 119.188 usec -> 2021-08-08T22:16:29.245.324.912 (XO: 17.245326347)

 - Block number: 3 starting at 2021-08-08T22:16:55.524.764.725 (XO: 43.524766957 - 43.525072958, t=37.2 degC)  Rank: 00087
   TM values: Max =   234  Min =  -241  Ave =    -4.50  Std =    22.16
   Min TM value:-241 at an offset of 119.425 usec -> 2021-08-08T22:16:55.524.884.150 (XO: 43.524886382)
   Max TM value:234 at an offset of 122.063 usec -> 2021-08-08T22:16:55.524.886.787 (XO: 43.524889020)
   Max dTM/dt [1/us]:1.96e+03 at an offset of 121.875 usec -> 2021-08-08T22:16:55.524.886.600 (XO: 43.524888832)
   Min dTM/dt [1/us]:-2.24e+03 at an offset of 119.275 usec -> 2021-08-08T22:16:55.524.884.000 (XO: 43.524886232)
