-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Dec  9 06:51:10 2021 UTC
-- File 20210807_220113_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.498976 to 21.999144
-each separately from 14s to 21s and from 498976us to 999144us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.499590 in 1 packets: 18
MIU packet time from 2021-08-07T22:01:30.498.976.000 to 2021-08-07T22:01:37.999.144.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.499590 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 16.2521994 to 23.2521762
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.9999768 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210807_220113_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.7511763 to 16.7514653
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.4995670  -- OK for SL --
   MIU time from 2021-08-07T22:01:30.498.976.000 to 2021-08-07T22:01:30.499.265.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.499590

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.2510323 to 24.2513213
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-07T22:01:37.998.855.000 to 2021-08-07T22:01:37.999.144.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.751176300 corrected by 40bit 10MHz counter of 16.252199400 obtained at 1Hz counter of 14. Result: 14.498978547
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-07T22:01:30.498.835.725
   Max 1st sample MIU time among all SL packets: 2021-08-07T22:01:37.999.003.725
   Min 1st sample XO time among all SL packets: 14.498838271
   Max 1st sample XO time among all SL packets: 21.999008122
   TM values from -255 to 225 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-07T22:01:30.498.835.725 (XO: 14.498838271 - 14.499144272, t=37.2 degC)  Rank: 00080
   TM values: Max =   223  Min =  -255  Ave =    -4.44  Std =    21.80
   Min TM value:-255 at an offset of 114.975 usec -> 2021-08-07T22:01:30.498.950.700 (XO: 14.498953247)
   Max TM value:223 at an offset of 114.675 usec -> 2021-08-07T22:01:30.498.950.400 (XO: 14.498952947)
   Max dTM/dt [1/us]:2.51e+03 at an offset of 117.313 usec -> 2021-08-07T22:01:30.498.953.037 (XO: 14.498955584)
   Min dTM/dt [1/us]:-2.57e+03 at an offset of 114.788 usec -> 2021-08-07T22:01:30.498.950.512 (XO: 14.498953059)

 - Block number: 2 starting at 2021-08-07T22:01:37.998.714.725 (XO: 21.998719121 - 21.999025122, t=37.2 degC)  Rank: 00082
   TM values: Max =   225  Min =  -228  Ave =    -4.42  Std =    24.08
   Min TM value:-228 at an offset of 126.763 usec -> 2021-08-07T22:01:37.998.841.487 (XO: 21.998845884)
   Max TM value:225 at an offset of 129.388 usec -> 2021-08-07T22:01:37.998.844.112 (XO: 21.998848509)
   Max dTM/dt [1/us]:1.77e+03 at an offset of 174.601 usec -> 2021-08-07T22:01:37.998.889.325 (XO: 21.998893721)
   Min dTM/dt [1/us]:-1.83e+03 at an offset of 126.600 usec -> 2021-08-07T22:01:37.998.841.325 (XO: 21.998845721)
