-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Dec  9 00:33:19 2021 UTC
-- File 20210807_102331_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.828320 to 24.988641
-each separately from 5s to 24s and from 828320us to 988641us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.159743 in 1 packets: 18
MIU packet time from 2021-08-07T10:23:39.828.320.000 to 2021-08-07T10:23:58.988.641.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.159743 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.9024838 to 25.9024199
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999361 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210807_102331_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.7308044 to 7.7310934
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.1596790  -- OK for SL --
   MIU time from 2021-08-07T10:23:39.828.320.000 to 2021-08-07T10:23:39.828.609.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.159743

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.8907724 to 26.8910614
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-07T10:23:58.988.352.000 to 2021-08-07T10:23:58.988.641.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.730804400 corrected by 40bit 10MHz counter of 6.902483800 obtained at 1Hz counter of 5. Result: 5.828323416
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-07T10:23:39.828.179.725
   Max 1st sample MIU time among all SL packets: 2021-08-07T10:23:58.988.500.725
   Min 1st sample XO time among all SL packets: 5.828183141
   Max 1st sample XO time among all SL packets: 24.988504487
   TM values from -258 to 225 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-07T10:23:39.828.179.725 (XO: 5.828183141 - 5.828489142, t=37.2 degC)  Rank: 00078
   TM values: Max =   225  Min =  -241  Ave =    -4.65  Std =    27.62
   Min TM value:-241 at an offset of 123.775 usec -> 2021-08-07T10:23:39.828.303.500 (XO: 5.828306916)
   Max TM value:225 at an offset of 171.513 usec -> 2021-08-07T10:23:39.828.351.238 (XO: 5.828354654)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 171.338 usec -> 2021-08-07T10:23:39.828.351.063 (XO: 5.828354479)
   Min dTM/dt [1/us]:-1.85e+03 at an offset of 123.525 usec -> 2021-08-07T10:23:39.828.303.250 (XO: 5.828306666)

 - Block number: 2 starting at 2021-08-07T10:23:58.988.211.725 (XO: 24.988215486 - 24.988521487, t=37.2 degC)  Rank: 00082
   TM values: Max =   212  Min =  -258  Ave =    -4.57  Std =    32.22
   Min TM value:-258 at an offset of 170.076 usec -> 2021-08-07T10:23:58.988.381.800 (XO: 24.988385562)
   Max TM value:212 at an offset of 113.813 usec -> 2021-08-07T10:23:58.988.325.537 (XO: 24.988329299)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 172.563 usec -> 2021-08-07T10:23:58.988.384.288 (XO: 24.988388049)
   Min dTM/dt [1/us]:-2.07e+03 at an offset of 156.088 usec -> 2021-08-07T10:23:58.988.367.813 (XO: 24.988371574)
