-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec  8 17:28:52 2021 UTC
-- File 20210807_021959_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.043311 to 52.823596
-each separately from 10s to 52s and from 43311us to 823596us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.439709 in 1 packets: 18
      21.099765 in 1 packets: 36
      12.239655 in 1 packets: 54
MIU packet time from 2021-08-07T02:20:12.043.311.000 to 2021-08-07T02:20:54.823.596.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.439709 in 1 packets: 18
      21.099765 in 1 packets: 36
      12.239655 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.9507393 to 53.9506000
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999701 in 1 packets: 18
      20.9999304 in 1 packets: 36
      11.9999602 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210807_021959_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.9940514 to 11.9943404
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.4396790  -- OK for SL --
   MIU time from 2021-08-07T02:20:12.043.311.000 to 2021-08-07T02:20:12.043.600.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.439709

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.4340194 to 21.4343084
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.0996950  -- OK for SL --
   MIU time from 2021-08-07T02:20:21.483.309.000 to 2021-08-07T02:20:21.483.598.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.099765

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.5340034 to 42.5342924
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.2396150  -- OK for SL --
   MIU time from 2021-08-07T02:20:42.583.363.000 to 2021-08-07T02:20:42.583.652.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.239655

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 54.7739074 to 54.7741964
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-07T02:20:54.823.307.000 to 2021-08-07T02:20:54.823.596.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 11.994051400 corrected by 40bit 10MHz counter of 11.950739300 obtained at 1Hz counter of 10. Result: 10.043312247
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-07T02:20:12.043.170.725
   Max 1st sample MIU time among all SL packets: 2021-08-07T02:20:54.823.455.725
   Min 1st sample XO time among all SL packets: 10.043171972
   Max 1st sample XO time among all SL packets: 52.823458842
   TM values from -274 to 219 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-07T02:20:12.043.170.725 (XO: 10.043171972 - 10.043477973, t=37.2 degC)  Rank: 00079
   TM values: Max =   210  Min =  -239  Ave =    -4.48  Std =    20.81
   Min TM value:-239 at an offset of 127.300 usec -> 2021-08-07T02:20:12.043.298.025 (XO: 10.043299272)
   Max TM value:210 at an offset of 127.013 usec -> 2021-08-07T02:20:12.043.297.737 (XO: 10.043298985)
   Max dTM/dt [1/us]:2.05e+03 at an offset of 129.688 usec -> 2021-08-07T02:20:12.043.300.412 (XO: 10.043301660)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 127.113 usec -> 2021-08-07T02:20:12.043.297.837 (XO: 10.043299085)

 - Block number: 2 starting at 2021-08-07T02:20:21.483.168.725 (XO: 19.483171319 - 19.483477320, t=37.2 degC)  Rank: 00078
   TM values: Max =   218  Min =  -264  Ave =    -4.54  Std =    27.64
   Min TM value:-264 at an offset of 159.801 usec -> 2021-08-07T02:20:21.483.328.525 (XO: 19.483331120)
   Max TM value:218 at an offset of 117.138 usec -> 2021-08-07T02:20:21.483.285.862 (XO: 19.483288457)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 119.763 usec -> 2021-08-07T02:20:21.483.288.487 (XO: 19.483291082)
   Min dTM/dt [1/us]:-2.49e+03 at an offset of 117.238 usec -> 2021-08-07T02:20:21.483.285.962 (XO: 19.483288557)

 - Block number: 3 starting at 2021-08-07T02:20:42.583.222.725 (XO: 40.583225250 - 40.583531251, t=37.2 degC)  Rank: 00077
   TM values: Max =   219  Min =  -249  Ave =    -4.50  Std =    17.24
   Min TM value:-249 at an offset of 97.088 usec -> 2021-08-07T02:20:42.583.319.812 (XO: 40.583322337)
   Max TM value:219 at an offset of 96.800 usec -> 2021-08-07T02:20:42.583.319.525 (XO: 40.583322050)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 99.425 usec -> 2021-08-07T02:20:42.583.322.150 (XO: 40.583324675)
   Min dTM/dt [1/us]:-2.54e+03 at an offset of 96.913 usec -> 2021-08-07T02:20:42.583.319.637 (XO: 40.583322162)

 - Block number: 4 starting at 2021-08-07T02:20:54.823.166.725 (XO: 52.823169841 - 52.823475842, t=37.2 degC)  Rank: 00081
   TM values: Max =   210  Min =  -274  Ave =    -4.48  Std =    27.60
   Min TM value:-274 at an offset of 187.988 usec -> 2021-08-07T02:20:54.823.354.713 (XO: 52.823357830)
   Max TM value:210 at an offset of 131.888 usec -> 2021-08-07T02:20:54.823.298.612 (XO: 52.823301729)
   Max dTM/dt [1/us]:2.03e+03 at an offset of 134.525 usec -> 2021-08-07T02:20:54.823.301.250 (XO: 52.823304367)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 132.013 usec -> 2021-08-07T02:20:54.823.298.737 (XO: 52.823301854)
