-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Dec  7 16:43:30 2021 UTC
-- File 20210805_142519_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.967807 to 18.018147
-each separately from 12s to 18s and from 17858us to 968096us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.049762 in 1 packets: 18
MIU packet time from 2021-08-05T14:25:34.967.807.000 to 2021-08-05T14:25:40.018.147.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.049762 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.8446859 to 19.8446660
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.9999801 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210805_142519_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      70 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.8124930 to 14.8127820
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.0497430  -- OK for SL --
   MIU time from 2021-08-05T14:25:34.967.807.000 to 2021-08-05T14:25:34.968.096.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.049762

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.8625250 to 19.8628140
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-05T14:25:40.017.858.000 to 2021-08-05T14:25:40.018.147.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.812493000 corrected by 40bit 10MHz counter of 13.844685900 obtained at 1Hz counter of 12. Result: 12.967810294
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-05T14:25:34.967.666.725
   Max 1st sample MIU time among all SL packets: 2021-08-05T14:25:40.018.006.725
   Min 1st sample XO time among all SL packets: 12.967670018
   Max 1st sample XO time among all SL packets: 18.018007786
   TM values from -278 to 266 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-05T14:25:34.967.666.725 (XO: 12.967670018 - 12.967976019, t=37.2 degC)  Rank: 00070
   TM values: Max =   266  Min =  -278  Ave =    -4.56  Std =    31.88
   Min TM value:-278 at an offset of 126.725 usec -> 2021-08-05T14:25:34.967.793.450 (XO: 12.967796744)
   Max TM value:266 at an offset of 126.438 usec -> 2021-08-05T14:25:34.967.793.162 (XO: 12.967796456)
   Max dTM/dt [1/us]:2.51e+03 at an offset of 124.413 usec -> 2021-08-05T14:25:34.967.791.137 (XO: 12.967794431)
   Min dTM/dt [1/us]:-2.54e+03 at an offset of 126.550 usec -> 2021-08-05T14:25:34.967.793.275 (XO: 12.967796569)

 - Block number: 2 starting at 2021-08-05T14:25:40.017.717.725 (XO: 18.017718785 - 18.018024786, t=37.2 degC)  Rank: 00085
   TM values: Max =   219  Min =  -266  Ave =    -4.46  Std =    30.55
   Min TM value:-266 at an offset of 186.438 usec -> 2021-08-05T14:25:40.017.904.163 (XO: 18.017905223)
   Max TM value:219 at an offset of 186.151 usec -> 2021-08-05T14:25:40.017.903.875 (XO: 18.017904936)
   Max dTM/dt [1/us]:2.12e+03 at an offset of 188.801 usec -> 2021-08-05T14:25:40.017.906.525 (XO: 18.017907586)
   Min dTM/dt [1/us]:-2.19e+03 at an offset of 126.250 usec -> 2021-08-05T14:25:40.017.843.975 (XO: 18.017845036)
