-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Dec  7 12:46:05 2021 UTC
-- File 20210805_114341_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.739087 to 37.427586
-each separately from 12s to 37s and from 168201us to 739376us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.428825 in 1 packets: 18
      12.258807 in 1 packets: 36
MIU packet time from 2021-08-05T11:43:56.739.087.000 to 2021-08-05T11:44:21.427.586.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.428825 in 1 packets: 18
      12.258807 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.9227398 to 38.9226574
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999572 in 1 packets: 18
      11.9999604 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210805_114341_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.6618275 to 14.6621165
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.4287830  -- OK for SL --
   MIU time from 2021-08-05T11:43:56.739.087.000 to 2021-08-05T11:43:56.739.376.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.428825

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.0908995 to 27.0911885
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.2587670  -- OK for SL --
   MIU time from 2021-08-05T11:44:09.168.201.000 to 2021-08-05T11:44:09.168.490.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.258807

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.3499555 to 39.3502445
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-05T11:44:21.427.297.000 to 2021-08-05T11:44:21.427.586.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.661827500 corrected by 40bit 10MHz counter of 13.922739800 obtained at 1Hz counter of 12. Result: 12.739090139
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-05T11:43:56.738.946.725
   Max 1st sample MIU time among all SL packets: 2021-08-05T11:44:21.427.445.725
   Min 1st sample XO time among all SL packets: 12.738949864
   Max 1st sample XO time among all SL packets: 37.427448236
   TM values from -266 to 208 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-05T11:43:56.738.946.725 (XO: 12.738949864 - 12.739255865, t=37.2 degC)  Rank: 00079
   TM values: Max =   208  Min =  -247  Ave =    -4.48  Std =    16.33
   Min TM value:-247 at an offset of 109.338 usec -> 2021-08-05T11:43:56.739.056.062 (XO: 12.739059201)
   Max TM value:208 at an offset of 111.875 usec -> 2021-08-05T11:43:56.739.058.600 (XO: 12.739061739)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 111.725 usec -> 2021-08-05T11:43:56.739.058.450 (XO: 12.739061589)
   Min dTM/dt [1/us]:-1.89e+03 at an offset of 109.150 usec -> 2021-08-05T11:43:56.739.055.875 (XO: 12.739059014)

 - Block number: 2 starting at 2021-08-05T11:44:09.168.060.725 (XO: 25.168062780 - 25.168368781, t=37.2 degC)  Rank: 00080
   TM values: Max =   188  Min =  -231  Ave =    -4.41  Std =    18.68
   Min TM value:-231 at an offset of 109.013 usec -> 2021-08-05T11:44:09.168.169.737 (XO: 25.168171792)
   Max TM value:188 at an offset of 124.275 usec -> 2021-08-05T11:44:09.168.185.000 (XO: 25.168187055)
   Max dTM/dt [1/us]:1.63e+03 at an offset of 124.100 usec -> 2021-08-05T11:44:09.168.184.825 (XO: 25.168186880)
   Min dTM/dt [1/us]:-1.83e+03 at an offset of 121.388 usec -> 2021-08-05T11:44:09.168.182.112 (XO: 25.168184168)

 - Block number: 3 starting at 2021-08-05T11:44:21.427.156.725 (XO: 37.427159235 - 37.427465236, t=37.2 degC)  Rank: 00081
   TM values: Max =   198  Min =  -266  Ave =    -4.40  Std =    27.58
   Min TM value:-266 at an offset of 109.900 usec -> 2021-08-05T11:44:21.427.266.625 (XO: 37.427269135)
   Max TM value:198 at an offset of 109.613 usec -> 2021-08-05T11:44:21.427.266.337 (XO: 37.427268847)
   Max dTM/dt [1/us]:1.55e+03 at an offset of 112.338 usec -> 2021-08-05T11:44:21.427.269.062 (XO: 37.427271572)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 185.076 usec -> 2021-08-05T11:44:21.427.341.800 (XO: 37.427344310)
