-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Dec  2 22:34:37 2021 UTC
-- File 20210801_102029_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.735824 to 16.396412
-each separately from 8s to 16s and from 396123us to 736113us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.660010 in 1 packets: 18
MIU packet time from 2021-08-01T10:20:40.735.824.000 to 2021-08-01T10:20:48.396.412.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.660010 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4295018 to 17.4294752
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999734 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210801_102029_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.1653266 to 10.1656156
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.6599830  -- OK for SL --
   MIU time from 2021-08-01T10:20:40.735.824.000 to 2021-08-01T10:20:40.736.113.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.660010

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.8255986 to 17.8258876
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-01T10:20:48.396.123.000 to 2021-08-01T10:20:48.396.412.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.165326600 corrected by 40bit 10MHz counter of 9.429501800 obtained at 1Hz counter of 8. Result: 8.735827302
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-01T10:20:40.735.683.725
   Max 1st sample MIU time among all SL packets: 2021-08-01T10:20:48.396.271.725
   Min 1st sample XO time among all SL packets: 8.735687026
   Max 1st sample XO time among all SL packets: 16.396273433
   TM values from -246 to 211 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-01T10:20:40.735.683.725 (XO: 8.735687026 - 8.735993027, t=37.2 degC)  Rank: 00077
   TM values: Max =   206  Min =  -246  Ave =    -4.44  Std =    22.89
   Min TM value:-246 at an offset of 127.425 usec -> 2021-08-01T10:20:40.735.811.150 (XO: 8.735814452)
   Max TM value:206 at an offset of 125.113 usec -> 2021-08-01T10:20:40.735.808.837 (XO: 8.735812139)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 124.988 usec -> 2021-08-01T10:20:40.735.808.712 (XO: 8.735812014)
   Min dTM/dt [1/us]:-1.84e+03 at an offset of 127.238 usec -> 2021-08-01T10:20:40.735.810.962 (XO: 8.735814264)

 - Block number: 2 starting at 2021-08-01T10:20:48.395.982.725 (XO: 16.395984432 - 16.396290433, t=37.2 degC)  Rank: 00082
   TM values: Max =   211  Min =  -238  Ave =    -4.46  Std =    20.47
   Min TM value:-238 at an offset of 130.125 usec -> 2021-08-01T10:20:48.396.112.850 (XO: 16.396114557)
   Max TM value:211 at an offset of 127.950 usec -> 2021-08-01T10:20:48.396.110.675 (XO: 16.396112382)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 127.838 usec -> 2021-08-01T10:20:48.396.110.562 (XO: 16.396112270)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 129.963 usec -> 2021-08-01T10:20:48.396.112.687 (XO: 16.396114395)
