-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Dec  2 09:03:34 2021 UTC
-- File 20210731_165637_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.245503 to 33.243761
-each separately from 9s to 33s and from 243472us to 985192us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.739111 in 1 packets: 18
      16.258280 in 1 packets: 36
MIU packet time from 2021-07-31T16:56:49.245.503.000 to 2021-07-31T16:57:13.243.761.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.739111 in 1 packets: 18
      16.258280 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.8238758 to 34.8237956
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.9999766 in 1 packets: 18
      16.9999432 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210731_165637_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.0693801 to 11.0696691
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.7390870  -- OK for SL --
   MIU time from 2021-07-31T16:56:49.245.503.000 to 2021-07-31T16:56:49.245.792.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.739111

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.8087561 to 18.8090451
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.2582230  -- OK for SL --
   MIU time from 2021-07-31T16:56:56.984.903.000 to 2021-07-31T16:56:56.985.192.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.258280

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.0672681 to 35.0675571
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-31T16:57:13.243.472.000 to 2021-07-31T16:57:13.243.761.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.069380100 corrected by 40bit 10MHz counter of 10.823875800 obtained at 1Hz counter of 9. Result: 9.245505135
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-31T16:56:49.245.362.725
   Max 1st sample MIU time among all SL packets: 2021-07-31T16:57:13.243.620.725
   Min 1st sample XO time among all SL packets: 9.245364859
   Max 1st sample XO time among all SL packets: 33.243622053
   TM values from -279 to 251 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-31T16:56:49.245.362.725 (XO: 9.245364859 - 9.245670860, t=37.2 degC)  Rank: 00078
   TM values: Max =   249  Min =  -279  Ave =    -4.54  Std =    23.52
   Min TM value:-279 at an offset of 103.238 usec -> 2021-07-31T16:56:49.245.465.962 (XO: 9.245468097)
   Max TM value:249 at an offset of 100.938 usec -> 2021-07-31T16:56:49.245.463.662 (XO: 9.245465797)
   Max dTM/dt [1/us]:2.26e+03 at an offset of 113.800 usec -> 2021-07-31T16:56:49.245.476.525 (XO: 9.245478660)
   Min dTM/dt [1/us]:-2.38e+03 at an offset of 115.925 usec -> 2021-07-31T16:56:49.245.478.650 (XO: 9.245480785)

 - Block number: 2 starting at 2021-07-31T16:56:56.984.762.725 (XO: 16.984766675 - 16.985072676, t=37.2 degC)  Rank: 00083
   TM values: Max =   251  Min =  -270  Ave =    -4.43  Std =    28.12
   Min TM value:-270 at an offset of 134.500 usec -> 2021-07-31T16:56:56.984.897.225 (XO: 16.984901175)
   Max TM value:251 at an offset of 137.150 usec -> 2021-07-31T16:56:56.984.899.875 (XO: 16.984903825)
   Max dTM/dt [1/us]:2.23e+03 at an offset of 137.038 usec -> 2021-07-31T16:56:56.984.899.762 (XO: 16.984903713)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 134.375 usec -> 2021-07-31T16:56:56.984.897.100 (XO: 16.984901050)

 - Block number: 3 starting at 2021-07-31T16:57:13.243.331.725 (XO: 33.243333052 - 33.243639053, t=37.2 degC)  Rank: 00081
   TM values: Max =   250  Min =  -254  Ave =    -4.46  Std =    21.39
   Min TM value:-254 at an offset of 127.025 usec -> 2021-07-31T16:57:13.243.458.750 (XO: 33.243460078)
   Max TM value:250 at an offset of 129.688 usec -> 2021-07-31T16:57:13.243.461.412 (XO: 33.243462740)
   Max dTM/dt [1/us]:1.83e+03 at an offset of 129.588 usec -> 2021-07-31T16:57:13.243.461.312 (XO: 33.243462640)
   Min dTM/dt [1/us]:-2.01e+03 at an offset of 126.875 usec -> 2021-07-31T16:57:13.243.458.600 (XO: 33.243459928)
