-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Dec  2 08:49:47 2021 UTC
-- File 20210731_165024_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.837446 to 17.936306
-each separately from 4s to 17s and from 837446us to 936306us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.098282 in 1 packets: 18
MIU packet time from 2021-07-31T16:50:31.837.446.000 to 2021-07-31T16:50:44.936.306.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.098282 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.9083067 to 18.9082632
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999565 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      3 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      3 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210731_165024_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.7457530 to 6.7460420
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.0982390  -- OK for SL --
   MIU time from 2021-07-31T16:50:31.837.446.000 to 2021-07-31T16:50:31.837.735.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.098282

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.8442810 to 19.8445700
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-31T16:50:44.936.017.000 to 2021-07-31T16:50:44.936.306.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.745753000 corrected by 40bit 10MHz counter of 5.908306700 obtained at 1Hz counter of 4. Result: 4.837449147
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-07-31T16:50:31.837.305.725
   Max 1st sample MIU time among all SL packets: 2021-07-31T16:50:44.936.165.725
   Min 1st sample XO time among all SL packets: 4.837308872
   Max 1st sample XO time among all SL packets: 17.936169708
   TM values from -281 to 239 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-31T16:50:31.837.305.725 (XO: 4.837308872 - 4.837614873, t=37.2 degC)  Rank: 00085
   TM values: Max =   239  Min =  -241  Ave =    -4.43  Std =    19.28
   Min TM value:-241 at an offset of 117.513 usec -> 2021-07-31T16:50:31.837.423.237 (XO: 4.837426385)
   Max TM value:239 at an offset of 120.188 usec -> 2021-07-31T16:50:31.837.425.912 (XO: 4.837429060)
   Max dTM/dt [1/us]:2.03e+03 at an offset of 120.038 usec -> 2021-07-31T16:50:31.837.425.762 (XO: 4.837428910)
   Min dTM/dt [1/us]:-1.83e+03 at an offset of 117.325 usec -> 2021-07-31T16:50:31.837.423.050 (XO: 4.837426197)

 - Block number: 2 starting at 2021-07-31T16:50:44.935.876.725 (XO: 17.935880707 - 17.936186708, t=37.2 degC)  Rank: 00082
   TM values: Max =   235  Min =  -281  Ave =    -4.40  Std =    22.14
   Min TM value:-281 at an offset of 69.450 usec -> 2021-07-31T16:50:44.935.946.175 (XO: 17.935950157)
   Max TM value:235 at an offset of 82.025 usec -> 2021-07-31T16:50:44.935.958.750 (XO: 17.935962732)
   Max dTM/dt [1/us]:2.39e+03 at an offset of 84.713 usec -> 2021-07-31T16:50:44.935.961.437 (XO: 17.935965420)
   Min dTM/dt [1/us]:-2.57e+03 at an offset of 82.163 usec -> 2021-07-31T16:50:44.935.958.887 (XO: 17.935962870)
