-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Nov 30 11:33:24 2021 UTC
-- File 20210729_115720_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.473198 to 75.190229
-each separately from 6s to 75s and from 189940us to 954951us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.076830 in 1 packets: 18
      19.883249 in 1 packets: 36
      21.792120 in 1 packets: 54
      4.728109 in 1 packets: 72
      13.234989 in 1 packets: 90
MIU packet time from 2021-07-29T11:57:29.473.198.000 to 2021-07-29T11:58:38.190.229.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.076830 in 1 packets: 18
      19.883249 in 1 packets: 36
      21.792120 in 1 packets: 54
      4.728109 in 1 packets: 72
      13.234989 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9360911 to 76.9358611
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999700 in 1 packets: 18
      19.9999333 in 1 packets: 36
      21.9999267 in 1 packets: 54
      3.9999867 in 1 packets: 72
      13.9999533 in 1 packets: 90
Temperature FPGA: 
      05 -> 48.5 degC in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      06 -> 47.2 degC in 18 packets: 91, 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Temperature XO: 
      C1 -> 37.2 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 89 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 89 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20210729_115720_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.4092903 to 8.4095793
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.0767990  -- OK for SL --
   MIU time from 2021-07-29T11:57:29.473.198.000 to 2021-07-29T11:57:29.473.487.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.076830

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.4863783 to 17.4866673
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.8831830  -- OK for SL --
   MIU time from 2021-07-29T11:57:38.550.317.000 to 2021-07-29T11:57:38.550.606.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.883249

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.3698503 to 37.3701393
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.7920470  -- OK for SL --
   MIU time from 2021-07-29T11:57:58.433.855.000 to 2021-07-29T11:57:58.434.144.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.792120

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 59.1621863 to 59.1624753
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.7280950  -- OK for SL --
   MIU time from 2021-07-29T11:58:20.226.264.000 to 2021-07-29T11:58:20.226.553.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.728109

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 63.8905703 to 63.8908593
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.2349430  -- OK for SL --
   MIU time from 2021-07-29T11:58:24.954.662.000 to 2021-07-29T11:58:24.954.951.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.234989

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 77.1258023 to 77.1260913
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-29T11:58:38.189.940.000 to 2021-07-29T11:58:38.190.229.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.409290300 corrected by 40bit 10MHz counter of 7.936091100 obtained at 1Hz counter of 6. Result: 6.473200762
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-29T11:57:29.473.057.725
   Max 1st sample MIU time among all SL packets: 2021-07-29T11:58:38.190.088.725
   Min 1st sample XO time among all SL packets: 6.473060486
   Max 1st sample XO time among all SL packets: 75.190090552
   TM values from -258 to 214 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-29T11:57:29.473.057.725 (XO: 6.473060486 - 6.473366487, t=37.2 degC)  Rank: 00077
   TM values: Max =   201  Min =  -258  Ave =    -4.50  Std =    24.39
   Min TM value:-258 at an offset of 165.051 usec -> 2021-07-29T11:57:29.473.222.775 (XO: 6.473225537)
   Max TM value:201 at an offset of 162.876 usec -> 2021-07-29T11:57:29.473.220.600 (XO: 6.473223362)
   Max dTM/dt [1/us]:1.86e+03 at an offset of 167.426 usec -> 2021-07-29T11:57:29.473.225.150 (XO: 6.473227912)
   Min dTM/dt [1/us]:-1.65e+03 at an offset of 164.938 usec -> 2021-07-29T11:57:29.473.222.663 (XO: 6.473225424)

 - Block number: 2 starting at 2021-07-29T11:57:38.550.176.725 (XO: 15.550178741 - 15.550484742, t=37.2 degC)  Rank: 00081
   TM values: Max =   182  Min =  -243  Ave =    -4.57  Std =    20.70
   Min TM value:-243 at an offset of 119.800 usec -> 2021-07-29T11:57:38.550.296.525 (XO: 15.550298541)
   Max TM value:182 at an offset of 122.438 usec -> 2021-07-29T11:57:38.550.299.162 (XO: 15.550301178)
   Max dTM/dt [1/us]:1.73e+03 at an offset of 122.263 usec -> 2021-07-29T11:57:38.550.298.987 (XO: 15.550301003)
   Min dTM/dt [1/us]:-1.47e+03 at an offset of 119.675 usec -> 2021-07-29T11:57:38.550.296.400 (XO: 15.550298416)

 - Block number: 3 starting at 2021-07-29T11:57:58.433.714.725 (XO: 35.433717056 - 35.434023057, t=37.2 degC)  Rank: 00077
   TM values: Max =   185  Min =  -236  Ave =    -4.48  Std =    17.63
   Min TM value:-236 at an offset of 84.050 usec -> 2021-07-29T11:57:58.433.798.775 (XO: 35.433801107)
   Max TM value:185 at an offset of 83.750 usec -> 2021-07-29T11:57:58.433.798.475 (XO: 35.433800807)
   Max dTM/dt [1/us]:1.58e+03 at an offset of 86.475 usec -> 2021-07-29T11:57:58.433.801.200 (XO: 35.433803532)
   Min dTM/dt [1/us]:-1.74e+03 at an offset of 83.875 usec -> 2021-07-29T11:57:58.433.798.600 (XO: 35.433800932)

 - Block number: 4 starting at 2021-07-29T11:58:20.226.123.725 (XO: 57.226125671 - 57.226431672, t=37.2 degC)  Rank: 00078
   TM values: Max =   214  Min =  -241  Ave =    -4.50  Std =    17.46
   Min TM value:-241 at an offset of 122.513 usec -> 2021-07-29T11:58:20.226.246.237 (XO: 57.226248184)
   Max TM value:214 at an offset of 120.175 usec -> 2021-07-29T11:58:20.226.243.900 (XO: 57.226245847)
   Max dTM/dt [1/us]:1.77e+03 at an offset of 120.050 usec -> 2021-07-29T11:58:20.226.243.775 (XO: 57.226245722)
   Min dTM/dt [1/us]:-1.75e+03 at an offset of 122.313 usec -> 2021-07-29T11:58:20.226.246.037 (XO: 57.226247984)

 - Block number: 5 starting at 2021-07-29T11:58:24.954.521.725 (XO: 61.954525375 - 61.954831376, t=37.2 degC)  Rank: 00080
   TM values: Max =   196  Min =  -232  Ave =    -4.44  Std =    22.02
   Min TM value:-232 at an offset of 104.725 usec -> 2021-07-29T11:58:24.954.626.450 (XO: 61.954630100)
   Max TM value:196 at an offset of 120.100 usec -> 2021-07-29T11:58:24.954.641.825 (XO: 61.954645475)
   Max dTM/dt [1/us]:1.96e+03 at an offset of 119.938 usec -> 2021-07-29T11:58:24.954.641.662 (XO: 61.954645313)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 117.375 usec -> 2021-07-29T11:58:24.954.639.100 (XO: 61.954642750)

 - Block number: 6 starting at 2021-07-29T11:58:38.189.799.725 (XO: 75.189801551 - 75.190107552, t=37.2 degC)  Rank: 00077
   TM values: Max =   187  Min =  -252  Ave =    -4.49  Std =    19.60
   Min TM value:-252 at an offset of 117.875 usec -> 2021-07-29T11:58:38.189.917.600 (XO: 75.189919427)
   Max TM value:187 at an offset of 117.538 usec -> 2021-07-29T11:58:38.189.917.262 (XO: 75.189919089)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 120.250 usec -> 2021-07-29T11:58:38.189.919.975 (XO: 75.189921802)
   Min dTM/dt [1/us]:-1.74e+03 at an offset of 117.650 usec -> 2021-07-29T11:58:38.189.917.375 (XO: 75.189919202)
