-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Nov 30 08:28:50 2021 UTC
-- File 20210729_092808_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.259816 to 56.702820
-each separately from 0s to 56s and from 241413us to 702820us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.981308 in 1 packets: 18
      6.179971 in 1 packets: 36
      22.280569 in 1 packets: 54
MIU packet time from 2021-07-29T09:28:11.259.816.000 to 2021-07-29T09:29:07.702.820.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.981308 in 1 packets: 18
      6.179971 in 1 packets: 36
      22.280569 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.8703637 to 57.8701774
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.9999070 in 1 packets: 18
      5.9999800 in 1 packets: 36
      21.9999267 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210729_092808_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.1301808 to 2.1304698
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 27.9812150  -- OK for SL --
   MIU time from 2021-07-29T09:28:11.259.816.000 to 2021-07-29T09:28:11.260.105.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 27.981308

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.1116848 to 30.1119738
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.1799510  -- OK for SL --
   MIU time from 2021-07-29T09:28:39.241.413.000 to 2021-07-29T09:28:39.241.702.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.179971

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.2919248 to 36.2922138
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.2804950  -- OK for SL --
   MIU time from 2021-07-29T09:28:45.421.673.000 to 2021-07-29T09:28:45.421.962.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.280569

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 58.5727088 to 58.5729978
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-29T09:29:07.702.531.000 to 2021-07-29T09:29:07.702.820.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.130180800 corrected by 40bit 10MHz counter of 1.870363700 obtained at 1Hz counter of 0. Result: 0.259817957
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-29T09:28:11.259.675.725
   Max 1st sample MIU time among all SL packets: 2021-07-29T09:29:07.702.679.725
   Min 1st sample XO time among all SL packets: 0.259677682
   Max 1st sample XO time among all SL packets: 56.702682444
   TM values from -225 to 197 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-29T09:28:11.259.675.725 (XO: 0.259677682 - 0.259983683, t=37.2 degC)  Rank: 00081
   TM values: Max =   159  Min =  -220  Ave =    -4.60  Std =    17.43
   Min TM value:-220 at an offset of 118.188 usec -> 2021-07-29T09:28:11.259.793.912 (XO: 0.259795870)
   Max TM value:159 at an offset of 120.825 usec -> 2021-07-29T09:28:11.259.796.550 (XO: 0.259798507)
   Max dTM/dt [1/us]:1.39e+03 at an offset of 120.638 usec -> 2021-07-29T09:28:11.259.796.362 (XO: 0.259798320)
   Min dTM/dt [1/us]:-1.48e+03 at an offset of 117.975 usec -> 2021-07-29T09:28:11.259.793.700 (XO: 0.259795657)

 - Block number: 2 starting at 2021-07-29T09:28:39.241.272.725 (XO: 28.241274621 - 28.241580622, t=37.2 degC)  Rank: 00077
   TM values: Max =   189  Min =  -219  Ave =    -4.50  Std =    24.84
   Min TM value:-219 at an offset of 150.425 usec -> 2021-07-29T09:28:39.241.423.150 (XO: 28.241425047)
   Max TM value:189 at an offset of 106.200 usec -> 2021-07-29T09:28:39.241.378.925 (XO: 28.241380822)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 108.913 usec -> 2021-07-29T09:28:39.241.381.637 (XO: 28.241383534)
   Min dTM/dt [1/us]:-1.71e+03 at an offset of 106.300 usec -> 2021-07-29T09:28:39.241.379.025 (XO: 28.241380922)

 - Block number: 3 starting at 2021-07-29T09:28:45.421.532.725 (XO: 34.421535216 - 34.421841217, t=37.2 degC)  Rank: 00075
   TM values: Max =   197  Min =  -225  Ave =    -4.53  Std =    18.67
   Min TM value:-225 at an offset of 123.075 usec -> 2021-07-29T09:28:45.421.655.800 (XO: 34.421658291)
   Max TM value:197 at an offset of 122.775 usec -> 2021-07-29T09:28:45.421.655.500 (XO: 34.421657991)
   Max dTM/dt [1/us]:1.86e+03 at an offset of 125.450 usec -> 2021-07-29T09:28:45.421.658.175 (XO: 34.421660666)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 122.875 usec -> 2021-07-29T09:28:45.421.655.600 (XO: 34.421658091)

 - Block number: 4 starting at 2021-07-29T09:29:07.702.390.725 (XO: 56.702393443 - 56.702699444, t=37.2 degC)  Rank: 00081
   TM values: Max =   168  Min =  -224  Ave =    -4.51  Std =    21.99
   Min TM value:-224 at an offset of 118.213 usec -> 2021-07-29T09:29:07.702.508.937 (XO: 56.702511656)
   Max TM value:168 at an offset of 104.775 usec -> 2021-07-29T09:29:07.702.495.500 (XO: 56.702498218)
   Max dTM/dt [1/us]:1.56e+03 at an offset of 120.638 usec -> 2021-07-29T09:29:07.702.511.362 (XO: 56.702514081)
   Min dTM/dt [1/us]:-1.68e+03 at an offset of 104.900 usec -> 2021-07-29T09:29:07.702.495.625 (XO: 56.702498343)
