-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Nov 29 13:11:50 2021 UTC
-- File 20210728_095043_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.923950 to 93.153435
-each separately from 4s to 93s and from 3997us to 924239us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.079758 in 1 packets: 18
      16.549541 in 1 packets: 36
      21.939625 in 1 packets: 54
      9.279582 in 1 packets: 72
      17.239451 in 1 packets: 90
      10.139505 in 1 packets: 108
MIU packet time from 2021-07-28T09:50:50.923.950.000 to 2021-07-28T09:52:19.153.435.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.079758 in 1 packets: 18
      16.549541 in 1 packets: 36
      21.939625 in 1 packets: 54
      9.279582 in 1 packets: 72
      17.239451 in 1 packets: 90
      10.139505 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.9081309 to 94.9078310
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999529 in 1 packets: 18
      15.9999461 in 1 packets: 36
      21.9999258 in 1 packets: 54
      8.9999696 in 1 packets: 72
      17.9999393 in 1 packets: 90
      9.9999664 in 1 packets: 108
Temperature FPGA: 
      05 -> 48.5 degC in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 109 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Number of generated alerts: 
      0 in 109 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20210728_095043_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.8320818 to 6.8323708
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.0797110  -- OK for SL --
   MIU time from 2021-07-28T09:50:50.923.950.000 to 2021-07-28T09:50:50.924.239.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.079758

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.9120818 to 19.9123708
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5494870  -- OK for SL --
   MIU time from 2021-07-28T09:51:04.003.997.000 to 2021-07-28T09:51:04.004.286.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.549541

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.4618578 to 36.4621468
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.9395510  -- OK for SL --
   MIU time from 2021-07-28T09:51:20.553.827.000 to 2021-07-28T09:51:20.554.116.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.939625

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 58.4016978 to 58.4019868
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.2795510  -- OK for SL --
   MIU time from 2021-07-28T09:51:42.493.741.000 to 2021-07-28T09:51:42.494.030.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.279582

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 67.6815378 to 67.6818268
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.2393910  -- OK for SL --
   MIU time from 2021-07-28T09:51:51.773.612.000 to 2021-07-28T09:51:51.773.901.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.239451

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 84.9212178 to 84.9215068
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.1394710  -- OK for SL --
   MIU time from 2021-07-28T09:52:09.013.352.000 to 2021-07-28T09:52:09.013.641.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.139505

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 95.0609778 to 95.0612668
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-28T09:52:19.153.146.000 to 2021-07-28T09:52:19.153.435.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.832081800 corrected by 40bit 10MHz counter of 5.908130900 obtained at 1Hz counter of 4. Result: 4.923954041
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-28T09:50:50.923.809.725
   Max 1st sample MIU time among all SL packets: 2021-07-28T09:52:19.153.294.725
   Min 1st sample XO time among all SL packets: 4.923813766
   Max 1st sample XO time among all SL packets: 93.153296046
   TM values from -265 to 232 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-28T09:50:50.923.809.725 (XO: 4.923813766 - 4.924119767, t=37.2 degC)  Rank: 00078
   TM values: Max =   232  Min =  -235  Ave =    -4.46  Std =    21.25
   Min TM value:-235 at an offset of 105.600 usec -> 2021-07-28T09:50:50.923.915.325 (XO: 4.923919366)
   Max TM value:232 at an offset of 103.475 usec -> 2021-07-28T09:50:50.923.913.200 (XO: 4.923917241)
   Max dTM/dt [1/us]:2.13e+03 at an offset of 103.363 usec -> 2021-07-28T09:50:50.923.913.087 (XO: 4.923917129)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 105.475 usec -> 2021-07-28T09:50:50.923.915.200 (XO: 4.923919241)

 - Block number: 2 starting at 2021-07-28T09:51:04.003.856.725 (XO: 18.003857738 - 18.004163739, t=37.2 degC)  Rank: 00078
   TM values: Max =   219  Min =  -242  Ave =    -4.41  Std =    25.62
   Min TM value:-242 at an offset of 117.175 usec -> 2021-07-28T09:51:04.003.973.900 (XO: 18.003974914)
   Max TM value:219 at an offset of 116.888 usec -> 2021-07-28T09:51:04.003.973.612 (XO: 18.003974626)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 114.800 usec -> 2021-07-28T09:51:04.003.971.525 (XO: 18.003972539)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 117.000 usec -> 2021-07-28T09:51:04.003.973.725 (XO: 18.003974739)

 - Block number: 3 starting at 2021-07-28T09:51:20.553.686.725 (XO: 34.553689508 - 34.553995509, t=37.2 degC)  Rank: 00075
   TM values: Max =   194  Min =  -245  Ave =    -4.55  Std =    20.69
   Min TM value:-245 at an offset of 119.400 usec -> 2021-07-28T09:51:20.553.806.125 (XO: 34.553808908)
   Max TM value:194 at an offset of 119.075 usec -> 2021-07-28T09:51:20.553.805.800 (XO: 34.553808583)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 121.725 usec -> 2021-07-28T09:51:20.553.808.450 (XO: 34.553811233)
   Min dTM/dt [1/us]:-2.01e+03 at an offset of 119.163 usec -> 2021-07-28T09:51:20.553.805.887 (XO: 34.553808670)

 - Block number: 4 starting at 2021-07-28T09:51:42.493.600.725 (XO: 56.493603503 - 56.493909504, t=37.2 degC)  Rank: 00072
   TM values: Max =   226  Min =  -246  Ave =    -4.58  Std =    21.77
   Min TM value:-246 at an offset of 107.988 usec -> 2021-07-28T09:51:42.493.708.712 (XO: 56.493711491)
   Max TM value:226 at an offset of 107.638 usec -> 2021-07-28T09:51:42.493.708.362 (XO: 56.493711141)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 110.275 usec -> 2021-07-28T09:51:42.493.711.000 (XO: 56.493713779)
   Min dTM/dt [1/us]:-2.29e+03 at an offset of 107.750 usec -> 2021-07-28T09:51:42.493.708.475 (XO: 56.493711254)

 - Block number: 5 starting at 2021-07-28T09:51:51.773.471.725 (XO: 65.773474777 - 65.773780778, t=37.2 degC)  Rank: 00079
   TM values: Max =   216  Min =  -238  Ave =    -4.48  Std =    21.39
   Min TM value:-238 at an offset of 119.363 usec -> 2021-07-28T09:51:51.773.591.087 (XO: 65.773594140)
   Max TM value:216 at an offset of 121.825 usec -> 2021-07-28T09:51:51.773.593.550 (XO: 65.773596603)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 121.650 usec -> 2021-07-28T09:51:51.773.593.375 (XO: 65.773596428)
   Min dTM/dt [1/us]:-2.09e+03 at an offset of 119.138 usec -> 2021-07-28T09:51:51.773.590.862 (XO: 65.773593915)

 - Block number: 6 starting at 2021-07-28T09:52:09.013.211.725 (XO: 83.013212969 - 83.013518970, t=37.2 degC)  Rank: 00078
   TM values: Max =   231  Min =  -265  Ave =    -4.43  Std =    27.51
   Min TM value:-265 at an offset of 137.475 usec -> 2021-07-28T09:52:09.013.349.200 (XO: 83.013350444)
   Max TM value:231 at an offset of 108.625 usec -> 2021-07-28T09:52:09.013.320.350 (XO: 83.013321594)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 139.813 usec -> 2021-07-28T09:52:09.013.351.537 (XO: 83.013352782)
   Min dTM/dt [1/us]:-2.50e+03 at an offset of 108.738 usec -> 2021-07-28T09:52:09.013.320.462 (XO: 83.013321706)

 - Block number: 7 starting at 2021-07-28T09:52:19.153.005.725 (XO: 93.153007045 - 93.153313046, t=37.2 degC)  Rank: 00083
   TM values: Max =   221  Min =  -261  Ave =    -4.40  Std =    29.09
   Min TM value:-261 at an offset of 108.100 usec -> 2021-07-28T09:52:19.153.113.825 (XO: 93.153115146)
   Max TM value:221 at an offset of 107.750 usec -> 2021-07-28T09:52:19.153.113.475 (XO: 93.153114796)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 110.388 usec -> 2021-07-28T09:52:19.153.116.112 (XO: 93.153117433)
   Min dTM/dt [1/us]:-2.25e+03 at an offset of 107.863 usec -> 2021-07-28T09:52:19.153.113.587 (XO: 93.153114908)
