-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Nov 29 12:06:56 2021 UTC
-- File 20210728_090107_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.804440 to 60.385303
-each separately from 3s to 60s and from 385014us to 924794us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.639791 in 1 packets: 18
      5.479696 in 1 packets: 36
      16.679911 in 1 packets: 54
      10.780020 in 1 packets: 72
MIU packet time from 2021-07-28T09:01:13.804.440.000 to 2021-07-28T09:02:10.385.303.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.639791 in 1 packets: 18
      5.479696 in 1 packets: 36
      16.679911 in 1 packets: 54
      10.780020 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.9254834 to 61.9252935
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.9999200 in 1 packets: 18
      4.9999833 in 1 packets: 36
      16.9999434 in 1 packets: 54
      10.9999634 in 1 packets: 72
Temperature FPGA: 
      03 -> 51.1 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      04 -> 49.8 degC in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 72 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 72 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210728_090107_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.7299240 to 5.7302130
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.6397110  -- OK for SL --
   MIU time from 2021-07-28T09:01:13.804.440.000 to 2021-07-28T09:01:13.804.729.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.639791

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.3699240 to 29.3702130
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.4796790  -- OK for SL --
   MIU time from 2021-07-28T09:01:37.444.520.000 to 2021-07-28T09:01:37.444.809.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.479696

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.8498920 to 34.8501810
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.6798550  -- OK for SL --
   MIU time from 2021-07-28T09:01:42.924.505.000 to 2021-07-28T09:01:42.924.794.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.679911

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 51.5300360 to 51.5303250
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.7799830  -- OK for SL --
   MIU time from 2021-07-28T09:01:59.604.705.000 to 2021-07-28T09:01:59.604.994.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.780020

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 62.3103080 to 62.3105970
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-28T09:02:10.385.014.000 to 2021-07-28T09:02:10.385.303.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.729924000 corrected by 40bit 10MHz counter of 4.925483400 obtained at 1Hz counter of 3. Result: 3.804443255
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-28T09:01:13.804.299.725
   Max 1st sample MIU time among all SL packets: 2021-07-28T09:02:10.385.162.725
   Min 1st sample XO time among all SL packets: 3.804302979
   Max 1st sample XO time among all SL packets: 60.385164496
   TM values from -242 to 206 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-28T09:01:13.804.299.725 (XO: 3.804302979 - 3.804608980, t=37.2 degC)  Rank: 00084
   TM values: Max =   206  Min =  -242  Ave =    -4.47  Std =    19.16
   Min TM value:-242 at an offset of 124.163 usec -> 2021-07-28T09:01:13.804.423.887 (XO: 3.804427142)
   Max TM value:206 at an offset of 123.863 usec -> 2021-07-28T09:01:13.804.423.587 (XO: 3.804426842)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 126.475 usec -> 2021-07-28T09:01:13.804.426.200 (XO: 3.804429455)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 123.963 usec -> 2021-07-28T09:01:13.804.423.687 (XO: 3.804426942)

 - Block number: 2 starting at 2021-07-28T09:01:37.444.379.725 (XO: 27.444381836 - 27.444687837, t=37.2 degC)  Rank: 00079
   TM values: Max =   174  Min =  -204  Ave =    -4.55  Std =    18.01
   Min TM value:-204 at an offset of 122.263 usec -> 2021-07-28T09:01:37.444.501.987 (XO: 27.444504099)
   Max TM value:174 at an offset of 124.950 usec -> 2021-07-28T09:01:37.444.504.675 (XO: 27.444506786)
   Max dTM/dt [1/us]:1.27e+03 at an offset of 124.763 usec -> 2021-07-28T09:01:37.444.504.487 (XO: 27.444506599)
   Min dTM/dt [1/us]:-1.43e+03 at an offset of 109.350 usec -> 2021-07-28T09:01:37.444.489.075 (XO: 27.444491186)

 - Block number: 3 starting at 2021-07-28T09:01:42.924.364.725 (XO: 32.924368075 - 32.924674076, t=37.2 degC)  Rank: 00078
   TM values: Max =   172  Min =  -241  Ave =    -4.46  Std =    30.12
   Min TM value:-241 at an offset of 183.026 usec -> 2021-07-28T09:01:42.924.547.750 (XO: 32.924551101)
   Max TM value:172 at an offset of 114.200 usec -> 2021-07-28T09:01:42.924.478.925 (XO: 32.924482276)
   Max dTM/dt [1/us]:1.67e+03 at an offset of 185.426 usec -> 2021-07-28T09:01:42.924.550.150 (XO: 32.924553501)
   Min dTM/dt [1/us]:-1.80e+03 at an offset of 196.176 usec -> 2021-07-28T09:01:42.924.560.900 (XO: 32.924564251)

 - Block number: 4 starting at 2021-07-28T09:01:59.604.564.725 (XO: 49.604567681 - 49.604873682, t=37.2 degC)  Rank: 00080
   TM values: Max =   165  Min =  -206  Ave =    -4.40  Std =    17.38
   Min TM value:-206 at an offset of 122.163 usec -> 2021-07-28T09:01:59.604.686.887 (XO: 49.604689843)
   Max TM value:165 at an offset of 124.775 usec -> 2021-07-28T09:01:59.604.689.500 (XO: 49.604692456)
   Max dTM/dt [1/us]:1.28e+03 at an offset of 124.638 usec -> 2021-07-28T09:01:59.604.689.362 (XO: 49.604692318)
   Min dTM/dt [1/us]:-1.29e+03 at an offset of 109.488 usec -> 2021-07-28T09:01:59.604.674.212 (XO: 49.604677168)

 - Block number: 5 starting at 2021-07-28T09:02:10.384.873.725 (XO: 60.384875495 - 60.385181496, t=37.2 degC)  Rank: 00080
   TM values: Max =   196  Min =  -224  Ave =    -4.51  Std =    18.66
   Min TM value:-224 at an offset of 111.250 usec -> 2021-07-28T09:02:10.384.984.975 (XO: 60.384986745)
   Max TM value:196 at an offset of 113.775 usec -> 2021-07-28T09:02:10.384.987.500 (XO: 60.384989270)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 113.625 usec -> 2021-07-28T09:02:10.384.987.350 (XO: 60.384989120)
   Min dTM/dt [1/us]:-1.95e+03 at an offset of 111.063 usec -> 2021-07-28T09:02:10.384.984.787 (XO: 60.384986558)
