-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Nov 29 09:33:11 2021 UTC
-- File 20210727_163414_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.391092 to 44.910678
-each separately from 1s to 44s and from 391092us to 910678us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      28.459246 in 1 packets: 18
      15.059473 in 1 packets: 36
MIU packet time from 2021-07-27T16:34:18.391.092.000 to 2021-07-27T16:35:01.910.678.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      28.459246 in 1 packets: 18
      15.059473 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.8337555 to 45.8336110
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.9999059 in 1 packets: 18
      14.9999496 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      06 -> 47.2 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210727_163414_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.2248485 to 3.2251375
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 28.4591510  -- OK for SL --
   MIU time from 2021-07-27T16:34:18.391.092.000 to 2021-07-27T16:34:18.391.381.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 28.459246

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.6842885 to 31.6845775
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0594230  -- OK for SL --
   MIU time from 2021-07-27T16:34:46.850.627.000 to 2021-07-27T16:34:46.850.916.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.059473

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.7440005 to 46.7442895
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-27T16:35:01.910.389.000 to 2021-07-27T16:35:01.910.678.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.224848500 corrected by 40bit 10MHz counter of 2.833755500 obtained at 1Hz counter of 1. Result: 1.391094330
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-07-27T16:34:18.390.951.725
   Max 1st sample MIU time among all SL packets: 2021-07-27T16:35:01.910.537.725
   Min 1st sample XO time among all SL packets: 1.390954054
   Max 1st sample XO time among all SL packets: 44.910541321
   TM values from -246 to 241 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-27T16:34:18.390.951.725 (XO: 1.390954054 - 1.391260055, t=37.2 degC)  Rank: 00080
   TM values: Max =   227  Min =  -246  Ave =    -4.34  Std =    19.13
   Min TM value:-246 at an offset of 129.125 usec -> 2021-07-27T16:34:18.391.080.850 (XO: 1.391083180)
   Max TM value:227 at an offset of 131.788 usec -> 2021-07-27T16:34:18.391.083.512 (XO: 1.391085842)
   Max dTM/dt [1/us]:1.85e+03 at an offset of 131.688 usec -> 2021-07-27T16:34:18.391.083.412 (XO: 1.391085742)
   Min dTM/dt [1/us]:-1.89e+03 at an offset of 128.925 usec -> 2021-07-27T16:34:18.391.080.650 (XO: 1.391082980)

 - Block number: 2 starting at 2021-07-27T16:34:46.850.486.725 (XO: 29.850489717 - 29.850795718, t=37.2 degC)  Rank: 00082
   TM values: Max =   237  Min =  -221  Ave =    -4.39  Std =    15.88
   Min TM value:-221 at an offset of 114.838 usec -> 2021-07-27T16:34:46.850.601.562 (XO: 29.850604555)
   Max TM value:237 at an offset of 117.563 usec -> 2021-07-27T16:34:46.850.604.287 (XO: 29.850607280)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 117.438 usec -> 2021-07-27T16:34:46.850.604.162 (XO: 29.850607155)
   Min dTM/dt [1/us]:-1.65e+03 at an offset of 114.688 usec -> 2021-07-27T16:34:46.850.601.412 (XO: 29.850604405)

 - Block number: 3 starting at 2021-07-27T16:35:01.910.248.725 (XO: 44.910252320 - 44.910558321, t=37.2 degC)  Rank: 00089
   TM values: Max =   241  Min =  -242  Ave =    -4.31  Std =    27.70
   Min TM value:-242 at an offset of 117.663 usec -> 2021-07-27T16:35:01.910.366.387 (XO: 44.910369983)
   Max TM value:241 at an offset of 117.375 usec -> 2021-07-27T16:35:01.910.366.100 (XO: 44.910369695)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 120.038 usec -> 2021-07-27T16:35:01.910.368.762 (XO: 44.910372358)
   Min dTM/dt [1/us]:-2.21e+03 at an offset of 117.488 usec -> 2021-07-27T16:35:01.910.366.212 (XO: 44.910369808)
