-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Nov 29 07:31:34 2021 UTC
-- File 20210727_153020_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.025852 to 18.407336
-each separately from 5s to 18s and from 25852us to 407336us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.380906 in 1 packets: 18
MIU packet time from 2021-07-27T15:30:28.025.852.000 to 2021-07-27T15:30:41.407.336.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.380906 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.9356544 to 19.9356110
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999566 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210727_153020_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.9615072 to 6.9617962
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.3808630  -- OK for SL --
   MIU time from 2021-07-27T15:30:28.025.852.000 to 2021-07-27T15:30:28.026.141.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.380906

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.3426592 to 20.3429482
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-27T15:30:41.407.047.000 to 2021-07-27T15:30:41.407.336.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 6.961507200 corrected by 40bit 10MHz counter of 6.935654400 obtained at 1Hz counter of 5. Result: 5.025852885
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-27T15:30:28.025.711.725
   Max 1st sample MIU time among all SL packets: 2021-07-27T15:30:41.407.195.725
   Min 1st sample XO time among all SL packets: 5.025712610
   Max 1st sample XO time among all SL packets: 18.407198309
   TM values from -265 to 237 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-27T15:30:28.025.711.725 (XO: 5.025712610 - 5.026018611, t=37.2 degC)  Rank: 00083
   TM values: Max =   237  Min =  -262  Ave =    -4.39  Std =    24.65
   Min TM value:-262 at an offset of 126.488 usec -> 2021-07-27T15:30:28.025.838.212 (XO: 5.025839098)
   Max TM value:237 at an offset of 126.175 usec -> 2021-07-27T15:30:28.025.837.900 (XO: 5.025838785)
   Max dTM/dt [1/us]:2.38e+03 at an offset of 128.788 usec -> 2021-07-27T15:30:28.025.840.512 (XO: 5.025841398)
   Min dTM/dt [1/us]:-2.51e+03 at an offset of 126.275 usec -> 2021-07-27T15:30:28.025.838.000 (XO: 5.025838885)

 - Block number: 2 starting at 2021-07-27T15:30:41.406.906.725 (XO: 18.406909308 - 18.407215310, t=37.2 degC)  Rank: 00080
   TM values: Max =   218  Min =  -265  Ave =    -4.48  Std =    20.04
   Min TM value:-265 at an offset of 115.263 usec -> 2021-07-27T15:30:41.407.021.987 (XO: 18.407024571)
   Max TM value:218 at an offset of 114.950 usec -> 2021-07-27T15:30:41.407.021.675 (XO: 18.407024259)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 117.650 usec -> 2021-07-27T15:30:41.407.024.375 (XO: 18.407026959)
   Min dTM/dt [1/us]:-2.15e+03 at an offset of 115.063 usec -> 2021-07-27T15:30:41.407.021.787 (XO: 18.407024371)
